Further to AIM 885 produces internally inconsistent results…
A new release, AIM885A appeared recently.
In the common theme of one step forward, two steps backwards, this version produces error popups when started.
The above popup appears twice when starting AIM885A. Just another symptom to undermine confidence in the system. It doesn’t make sense to me, and the program appears to otherwise start and run. Continue reading AIM 885A produces internally inconsistent results
This article builds on Accuracy of AIMuhf system – AIM865A vs AIM882 on a ferrite cored inductor by adding a measurement using AIM885 of the same test inductor.
Above is the test inductor. It comprises 6 turns of 0.5mm PVC insulated wire wound on a BN-43-202 binocular balun core. Continue reading Accuracy of AIMuhf system – AIM865A vs AIM882 vs AIM885 on a ferrite cored inductor
New features in version 885 is
Fixed “Refer to Antenna” function… so lets check it out.
The AIMuhf was calibrated using the supplied OSL components and AIM885.
Above is a sweep of a 1.05m length of Belden 8262 (RG58C/U) cable (with o/c termination) attached to the AIMuhf. Continue reading AIM885 Refer To Antenna checkout
In other posts, I have commented on the apparent inconsistency of AIMuhf measurements.
One of the devices I often wish to measure is a ferrite cored choke such as those used for a Guanella 1:1 balun. A small test inductor was made to provide a common device for measurement across instruments, and versions of software, though small, it has similar characteristics to the larger inductors more commonly measured.
Above is the test inductor. It comprises 6 turns of 0.5mm PVC insulated wire wound on a BN-43-202 binocular balun core. Continue reading Accuracy of AIMuhf system – AIM865A vs AIM882 on a ferrite cored inductor
At Where is the best place to measure feed point VSWR I discussed location of the VSWR meter and projection of its reading to another point on a known transmission line.
A correspondent has taken me to task and citing Telepost’s LP-100A manual: Continue reading LP-100A manual advice on VSWR measurement
Jaycar have sold the MP3090 (Manson SPS-9400) over many years. I have one dated 2001 and they are still in their catalog in 2015. Manson is a Chinese manufacturer of medium quality power supplies, at reasonable prices. Continue reading Jaycar MP3090 / Manson SPS-9400 repair
At Diamond X-50N #2 at VK2OMD I reported some details of a new antenna at VK2OMD, including some baseline measurements using a Rigexpert AA-600.
This article expands that information with some measurements made using AIMuhf, and gives some contrasts between the two tools.
The AA-600 measurements were made 26/11/2014, and AIMuhf a couple of months later on 29/01/2015. The system measured should have been identical, the feed line is 9.83m of LDF4-50A with N-type connectors at both ends. Continue reading Diamond X50N measurements – AIMuhf and AA-600
I have spent considerable time over the last few weeks trying to salvage something useful and reliable from the wreckage that is AIMuhf and its PC client program AIM. Continue reading AIMuhf – my strategy