## Calculation of impedance of a ferrite toroidal inductor – from first principles

A toroidal inductor is a resonator, though it can be approximated as a simple inductor at frequences well below its self resonant frequency (SRF). Lets take a simple example, a ferrite toroid of rectangular cross section.

From the basic definition $$\mu=B/H$$ we can derive the relationship that the flux density in the core with current I flowing through N turns is given by $$B=\frac{\mu_0 \mu_r N I}{2 \pi r}$$. Continue reading Calculation of impedance of a ferrite toroidal inductor – from first principles

## nanoVNA – measuring cable velocity factor – demonstration

The article nanoVNA – measuring cable velocity factor discussed ways of measuring the velocity factor of common coax cable. This article is a demonstration of one of the methods, 2: measure velocity factor with your nanoVNA then cut the cable.

Two lengths of the same cable were selected to measure with the nanoVNA and calculate using Velocity factor solver. The cables are actually patch cables of nominally 1m and 2.5m length. Importantly they are identical in EVERY respect except the length, same cable off the same roll, same connectors, same temperature etc.

Above is the test setup. The nanoVNA is OSL calibrated at the external side of the SMA saver (the gold coloured thing on the SMA port), then an SMA(M)-N(F) adapter and the test cable. The other end of the test cable is left open (which is fine for N type male connectors). Continue reading nanoVNA – measuring cable velocity factor – demonstration

## nanoVNA – measuring cable velocity factor

With the popularity of the nanoVNA, one of the applications that is coming up regularly in online discussion is the use to measure velocity factor of cable and / or tuning of phasing sections in antenna feeds.

## ‘Tuning’ electrical lengths of transmission line sections

Online experts offer a range of advice including:

1. use the datasheet velocity factor;
2. measure velocity factor with your nanoVNA then cut the cable;
3. measure the ‘tuned’ length observing input impedance of the section with the nanoVNA; and
4. measure the ‘tuned’ length using the nanoVNA TDR facility.

All of these have advantages and pitfalls in some ways, some are better suited to some applications, others may be quite unsuitable.

Let’s make the point that these sections are often not highly critical in length, especially considering that in actual use, the loads are not perfect. One application where they are quite critical is the tuned interconnections in a typical repeater duplexer where the best response depends on quite exact tuning of lengths. Continue reading nanoVNA – measuring cable velocity factor

## PllLdr application – ATTiny44 & AD9833 in Codan 6801

The Codan 6801 is an older SSB transceiver using a single crystal per simplex SSB channel, for up to 10 channels. The channel switch selects the crystal and also a band pass filter for that channel.

The cost of crystals to populate the 6801 runs towards $1000. A recent project implemented a functional replacement for the crystals using PllLdr and an inexpensive DDS module.The cost of crystals to populate the 6801 runs towards$1000. A recent project implemented a functional replacement for the crystals using PllLdr and an inexpensive DDS module suitable for use in the ham bands.

Above, the modified radio with 8 channels on ham bands (this radio is missing the last two channel filters, so it is only equipped internally for 8 channels). Continue reading PllLdr application – ATTiny44 & AD9833 in Codan 6801

## Small untuned loop for receiving – it’s not rocket science

I have written several articles on untuned loops for receiving, as have others. A diversity of opinions abounds over several aspects, but opinions don’t often translate to sound theory.

This article analyses a simple untuned / unmatched loop in the context of a linear receive system.

## An example simple loop for discussion

Let’s consider a simple single turn untuned loop with an ideal broadband transformer. The example loop is 3.14m perimeter and 10mm diameter conductor situated in free space. The loop has perimeter 0.0744λ at 7.1MHz, less than λ/10 up to 9MHz, so we can regard that loop current is uniform in magnitude and phase. This simplifies analysis greatly.

Above is a schematic diagram of the example loop. The transformer initially is a 1:1 ideal transformer, it serves to isolate the loop from a coaxial feedline, allowing fairly good loop symmetry and reduction of common mode feed line current contribution to pickup. This works, and subject to symmetry and a good transformer design, it will work well over the stated frequency range, though its gain at some frequencies might not be sufficient to overcome receiver internal noise. Continue reading Small untuned loop for receiving – it’s not rocket science

## YouLoop-2T and the self resonance bogey at MF/lowHF

Small untuned loop for receiving – simple model with transformer gave a simple model for analysing a loop and and Towards understanding the YouLoop-2T at MF/lowHF  applied that to the YouLoop-2T.

Above is the Airspy Youloup-2T. Try to put the two turns thing out of your mind, it is misleading, panders to some common misunderstanding, and so does not help understanding.

It would seem that many are quite confused by information from Airspy. The following quote from an online forum captures the confusion. Continue reading YouLoop-2T and the self resonance bogey at MF/lowHF

## (How) does this balun work? (Tonna / F9FT balun for Yagis)

This article proposes an explanation of how the balun used on some Tonna Yagis works.

It appears Tonna no longer manufactures these antennas, I do not know if this design is novel. I do not recall seeing them used by other manufacturers, they may be protected by patent.

Above is a pic of the balun structure on a 2m antenna.

Above, the manual shows that the black sleeve on the balun sleeve would be slid up over the coax connector, making a neat finish. There are slightly different versions for 70cm antennas. Continue reading (How) does this balun work? (Tonna / F9FT balun for Yagis)