## Quality of termination used for calibration

Some of us use a resistor as a load for testing a transmitter or other RF source. In this application they are often rated for quite high power and commonly called a dummy load. In that role, they usually do not need to be of highly accurate impedance, and commercial dummy loads will often be specified to have maximum VSWR in the range 1.1 to 1.5 (Return Loss (RL) from 26 to 14dB) over a specified frequency range.

We also use a known value resistor for measurement purposes, and often relatively low power rating but higher impedance accuracy. They are commonly caused terminations, and will often be specified to have maximum VSWR in the range 1.01 to 1.1 (RL from 46 to 26dB) over a specified frequency range.

## Return Loss

It is more logical to discuss this subject in terms of Return Loss rather than VSWR.

Return Loss is defined as the ratio of incident to reflected power at a reference plane of a network. It is expressed in dB as 20*log(Vfwd/Vref). Continue reading Exploiting your antenna analyser #30

## Resolving the sign of reactance – a method – Smith chart detail

Exploiting your antenna analyser #28 gave an example of use of one method to resolve the sign of reactance comparing measurements made with a slightly longer known transmission line.

One way to predict the input impedance to the longer line is using a Smith chart. This article presents a Smith chart prediction of the expected input impedance of a 8′ section of RG8 at 14.17Mhz (vf=0.66, length=0.175λ) for the cases of Zload being 60.3+j26.9Ω and  60.3-j26.9Ω.

## 60.3+j26.9Ω

The impedance is normalised to 50Ω and plotted on the Smith chart, point 1 above. A radial from the centre through point 1 is drawn to the edge of the chart. Another radial is drawn a distance towards the generator of 0.175λ and using a pair of dividers or ruler, point 2 is plotted on that radial at the same distance from the centre (same VSWR) as point 1.

These points are on a constant VSWR arc but the arc has not been draw because the two arcs would overlap and might be confusing to some readers.  Continue reading Exploiting your antenna analyser #29

## Resolving the sign of reactance – a method

Many analysers do not measure the sign of reactance, and display the magnitude of reactance, and likewise for magnitude of phase and magnitude of impedance… though they are often incorrectly and misleadingly labelled otherwise.

The article The sign of reactance explains the problem and dismisses common recipes for resolving the sign of reactance as not general and not reliable.

This article gives an example of one method that may be useful for resolving the sign of reactance.

My correspondent has measured VSWR=1.68 and |Z|=66 and needs to know R and X. From those values we can calculate R=60.3 and |X|=26.9.

## Method

The method involves adding a short series section of known line, short enough to provide a measurement difference in R, and that R would be different for the case of =ve and -ve X, all of these measured at the same frequency. Continue reading Exploiting your antenna analyser #28

## An Insertion VSWR test gone wrong

We often learn more from failures than successes, this exercise is one of those opportunities.

An online poster tried to validate his newly purchased MFJ-918 by measuring Insertion VSWR.

That is done preferably by measuring a good termination (dummy load) to validate that it has a very low VSWR, then inserting the Device Under Test (DUT) and measuring the VSWR as a result of insertion of the DUT.

The poster did not mention measurement of the dummy load alone, and it is a type that warrants validation.

Above is the poster’s test setup, his Rigexpert AA-170 is connected to the balun’s input jack using a M-M adapter. The output wires on the balun form a rough circle of about 550mm perimeter by eye. Continue reading Exploiting your antenna analyser #27

## Find coax cable velocity factor using a very basic analyser

A common task is to measure the velocity factor of a sample of coaxial transmission line using an instrument that lacks facility to backout cable sections or measure SOL calibration (as discussed in other articles in this series). The older models and newer budget models often fall into this category.

The manuals for such instruments often explain how to measure coaxial cable velocity factor, and the method assumes there is zero offset at the measurement terminals (whether they be the built-in terminals or some fixture / adapters). In fact even the connectors are a source of error, especially UHF series connectors.

It is the failure to read exactly Z=0+j0Ω with a S/C applied to the measurement terminals that adversely impacts efforts to measure resonant frequency of a test line section.

The method described here approximately nulls out offsets in the instrument, measurement fixture, and even in the connectors used and for that reason may sometimes be of use with more sophisticated analysers.

## Exploiting your antenna analyser – contents

Exploiting your antenna analyser #30 Quality of termination used for calibration

Exploiting your antenna analyser #29 Resolving the sign of reactance – a method – Smith chart detail

Exploiting your antenna analyser #28 Resolving the sign of reactance – a method

Exploiting your antenna analyser #27 An Insertion VSWR test gone wrong

Exploiting your antenna analyser #26 Find coax cable velocity factor using a very basic analyser

Exploiting your antenna analyser #25 Find coax cable velocity factor using an antenna analyser without using SOL calibration

Exploiting your antenna analyser #24 Find coax cable velocity factor using an antenna analyser with SOL calibration

Exploiting your antenna analyser #23 Seeing recent discussion by online experts insisting that power relays are not suitable to RF prompts an interesting and relevant application of a good antenna analyser Continue reading Exploiting your antenna analyser – contents

## Find coax cable velocity factor using an antenna analyser without using SOL calibration

A common task is to measure the velocity factor of a sample of coaxial transmission line using an instrument without using SOL calibration.

Whilst this seems a trivial task with a modern antenna analyser, it seems to challenge many hams.

We will use a little test fixture that I made for measuring small components, and for which I have made test loads for SOL calibration. We will find the frequency where reactance passes through zero at the first parallel resonance of an O/C stub section, this is at a length of approximately λ/2 (a good approximation for low loss coaxial cables above about 10MHz).

We will use a little test fixture that I made for measuring small components, and for which I have made test loads for SOL calibration.

The text fixture used for this demonstration is constructed on a SMA(M) PCB connector using some machined pin connector strip and N(M)-SMA(F) adapters to connect to the instrument.

Above is a pic of the test fixture with adapters (in this case on a AA-600). Continue reading Exploiting your antenna analyser #25

## Find coax cable velocity factor using an antenna analyser with SOL calibration

A common task is to measure the velocity factor of a sample of coaxial transmission line using an instrument that supports SOL calibration, an AIMuhf in this example.

Whilst this seems a trivial task with a modern antenna analyser, it seems to challenge many hams.

There are a thousand recipes, I am going to demonstrate just one that suits the instrument and application.

We will use a little test fixture that I made for measuring small components, and for which I have made test loads for SOL calibration. We will find the frequency where reactance passes through zero at the first parallel resonance of an O/C stub section, this is at a length of approximately λ/2 (a good approximation for low loss coaxial cables above about 10MHz).

The text fixture used for this demonstration is constructed on a SMA(M) PCB connector using some machined pin connector strip and N(M)-SMA(F) adapters to connect to the instrument.

Above is a pic of the test fixture with adapters (in this case on a AA-600). Continue reading Exploiting your antenna analyser #24

## Exploiting your antenna analyser #23

Seeing recent discussion by online experts insisting that power relays are not suitable to RF prompts an interesting and relevant application of a good antenna analyser.

Above is a sweep of an A/B changeover relay intended for HF application at up to 100W and lowish VSWR. The sweep is actually from 1 to 61MHz to be confident that there is not poor behaviour just outside of the HF range that might present on another implementation of the same design. Continue reading Exploiting your antenna analyser #23

## Exploiting your antenna analyser #22

On a transmission line with standing waves, the voltage varies cyclically along the line, and is dependent also on power.

This article explains a method to use an analyser to predict the peak voltage level at a point for a given frequency and power based on measurement or estimation of complex Z or Y at that point using a suitable antenna analyser.

## The problem

Lets say you have some critical  voltage breakdown limit and want to use your analyser to find any non-compliance at the proposed power level.

Let us assume that the not-to-exceed voltage at that point is 1000Vpk. Let’s allow a little margin for variation due to factors not fixed, let’s actually use 800Vpk as the limit. We will use the maximum permitted power in Australia, 400W.