A new release, AIM914 appeared recently.
In the common theme of one step forward, two steps backwards, this version has defects that were not present in AIM910B.
Let’s review the internal consistency of this part of the display screen.
Most of the values given above are calculated from a single measurement value, and should be internally consistent. That measurement value is translated to different quantities, many based on the chosen Zref (75Ω in this case). Continue reading AIM 914 produces internally inconsistent results
Given that most versions of the AIM software that I have tried have had serious defects, I approach the latest release, AIM912, with caution.
An interesting opportunity presented when a correspondent sent me some .scn files captured from an AIM4170B using AIM912.
Above is the correspondents .scn file opened in AIM912. Continue reading AIM system – AIM912 initial checkout
The ‘unknown’ toroid is wound with a single turn and measured with a VNA, an AIMuhf in this case.
Of interest in the first instance is the apparent inductance of the single turn winding at low frequencies where typically permeability µ is fairly constant and core loss is fairly low. Continue reading Characterising an unknown ferrite toroid
Yet another release of AIM software is available, 910A at the time of writing. I have downloaded and tested 8 versions this year, most have been wanting. Again, there is very little detail on what has changed and likely impact on historical measurments.
A quick set of measurements was made on my test inductor pictured above. Continue reading Accuracy of AIMuhf system – AIM910A vs several recent versions on a ferrite cored inductor
A lost soul searching for enlightenment on impedance transformation sought advice on a transformer at 2.4 : 1 BALUN.
Inevitably one of the forum experts counselled:
Assuming your quad is a single-band HF antenna, a conventional transformer using #2 powdered iron would be my choice for the balun function. The reactance of the secondary winding would need to be at least 600 ohms.
So, let’s put the forum expert’s advice to a practical test.
Fleshing out the proposed solution
I have at hand a T200-2 core, so lets calculate the secondary turns to satisfy the proposed solution.
Above is calculation from a popular online calculator. For 14MHz, the secondary should be at least 23.8t. We will use 24t. Continue reading 2.4:1 balun design failure
I mentioned at A walk through of a practical application of AIMuhf/AIM900 that I wasn’t all together happy with feed point R at resonance, at 40Ω it was perhaps a touch high for a 2m quarter wave ground plane on a largish vehicle roof.
Repeated measurement of the DC resistance from the coax plug sheild to car body yielded unstable resistance ranging from 1 to 10Ω. If stable low DC resistance is not achieved, this feed line won’t work properly for RF. Continue reading A walk through of a practical application of AIMuhf/AIM900A #2
This article describes the use of the Array Solutions AIMuhf/AIM900 to test a mobile antenna installation, a quarter wave whip for 2m with about 4m of RG58 cable which has been previously installed and tuned.
The exercise is motivated by a perception that the antenna is not working as well as it should.
Above is a scan of the VSWR. It indicates problems, there should be a main VSWR dip around the high end of the 2m band (147MHz), but instead the minimum is nearer 160MHz. Clearly there is an antenna connected to the far end of the line in some form (ie the inner conductor is not simply broken), but there could be a high resistance in the inner conductor or shield connection (the latter is common issue with this type of antenna base). Continue reading A walk through of a practical application of AIMuhf/AIM900
At Accuracy of AIMuhf system – AIM865A vs AIM882 vs AIM885 on a ferrite cored inductor I compared measurement of a small ferrite cored inductor on three ‘production’ releases of the AIM program.
Since then, AIM885A was released and it had problems as mentioned at AIM 885A produces internally inconsistent results.
Since then, AIM882, AIM885, and AIM885A have been pulled.
The developer advised on his support forum
I’m sorry but version 885A has some problems. I’m working an update. In the meantime use the mature version AIM_865A.
Today, he posted a link to AIM885Gx (a declared beta release which presumably fixes problems in the pulled versions AIM882, AIM885, and AIM885A). Continue reading Accuracy of AIMuhf system – AIM865A vs AIM885Gx on a ferrite cored inductor
Further to AIM 885 produces internally inconsistent results…
A new release, AIM885A appeared recently.
In the common theme of one step forward, two steps backwards, this version produces error popups when started.
The above popup appears twice when starting AIM885A. Just another symptom to undermine confidence in the system. It doesn’t make sense to me, and the program appears to otherwise start and run. Continue reading AIM 885A produces internally inconsistent results