G4YDM described his balun at Ham Radio – What Is a Balun and How to Make One Cheaply.
With a title like that it is sure to have wide appeal, but it isn’t anything too novel, it is simply an air solenoid of 50Ω coax cable as a common mode choke, commonly known as an Ugly Balun.
He gives some instructions for one of several constructions:
When wrapping your coax around the pipe don’t use too much force as it may damage the inner braid and space the turns away from each other by a millimetre or two. R-G-2-1-3 coax around 21 feet used with 5 inch pipe will handle 400 watts pf power.
Above is a pic of the third construction which appears to be 21′ of RG213 on a 5″ PVC former:
He gives some performance measurements adjacent to the pic above:
Using a dummy load connected to the choke and transmitting 100 watts from my transmitter indicated an S.W.R. readings of around 1.5 to 1 at 3.5 Megahertz when testing 28 Megahertz the S.W.R. reading came down to 1.1 to 1 which is an excellent match. …
The test described above seems to simply be a dummy load connected to one end of 21′ of RG213 and the transmitter with VSWR meter feeding the other end. To be meaningful we need to know the impedance of the dummy load, indeed to be meaningful it needs to be 50Ω, so lets assume that is the case. Continue reading G4YDM balun
Remote speaker-microphones and DMR portables discussed RF ingress to Speaker Mics (RSM) used with DMR radios in digital mode.
I purchased a RSM branded Kenwood but obviously a Chinese fake for an MD-390 for about $5 posted, but it turned out to be lousy with RF interference in the form of the ‘motorboat noise’ on transmit audio.
Dismantling the RSM I found there is precious little RF filtering, just a single SMD cap near the electret capsule.
Above is the modified RSM. Continue reading Another speaker mic modification to reduce RF interference
At Rigexpert Antscope v4.3.1 released I commented on a new release of Antscope.
Correspondents have asked where I obtained v4.3.1.
Well, it seems the Rigexpert website is broken again, the URL to list the Antscope downloads produces garbage. Nevertheless, you can get a directory listing at https://www.rigexpert.com/files/antscope/ and yes, you will note that v4.3.1 is not listed… so it seems to have been either pulled due to defects or it is just a consequence of the web site problems.
Little loss, I use v4.2.57 on Rigexpert’s advice as it has better scales for impedance plots… and v4.2.57 is still published (at the time or writing) https://www.rigexpert.com/files/antscope/antscope040257.zip .
A correspondent asked about the use of a Jaycar LO1238 ferrite core in VK3IL’s EFHW matching unit for 40m and up. The LO1238 implementation would use 3t primary and 24t secondary on the core.
If the transformer is simply used without an ATU between it and the radio, and we assume that the antenna system is adjusted to present low VSWR(50) to the radio, a simple approximation involves calculating the magnetising admittance of the 3t 50Ω winding, and calculating the portion of total input power that is dissipated in that admittance.
Using the calculator at Calculate ferrite cored inductor, the admittance (G+jB) of the 3t winding is 0.00177-j0.00204S. (The impedance of a sample wind could be measured with a suitable analyser and converted to admittance.) Continue reading End fed matching – VK3IL design on LO1238
A correspondent wrote seeking clarification of the Telepost LP-100A claims re impedance measurement in the context of some of my previous articles on the sign of reactance.
I could see several mentions in the LP-100A manual and the LP_100Plot documentation and they do seem a little inconsistent.
The LP-100A manual states very clearly:
Note: The LP-100A cannot determine the sign of X automatically.
If you QSY up from your current frequency, and the reactance goes up, then the reactance is inductive (sign is “+”), and conversely if it goes down, then the reactance is capacitive (sign is “-“). A suitable distance is QSY is about 100 kHz or more. The LP-Plot program has the ability to determine sign automatically, since it can control your transmitter’s frequency. When it plots a range of frequencies, it uses the slope of the reactance curve to determine sign, and plots the results accordingly.
The first part states clearly that the instrument cannot directly measure the sign of reactance, and presumably measures the magnitude of reactance |X|.
Lets explore the second part in light of the overarching statement of the first part.
Above is the calculated R and X looking into 7m of Belden RG58C/U with a load 25+j0Ω. Also shown is |X|(as would be measured by the LP-100A) and calculated magnitude of phase of R,X, |φ|. Continue reading LP-100A impedance measurement
It seems yet another new version of Rigexpert Antscope has been released, and it maintains the scale limits available for R,X plots to +/-2000Ω, it still does not allow the range permitted by v4.2.57 (+/-5000Ω).
No change details provided by Rigexpert.
Back to v4.2.57, though it is very likely it has undisclosed defects fixed in later releases.
Bottom line is that if you want an analyser with direct graphing of impedances over 2000Ω (eg measuring common mode choke impedance), think of a different analyser.
Recent advertising of the Boafeng DM-5 DMR portable prompted a review of available products at the low end of the market.
The Boafeng DM-5 was dismissed on a desk study that revealed that advertisements were misleading and deceptive in that the claimed Tier 2 support was not yet delivered and was not included in the advertised price. Previous experience with Boafeng also factored against that solution.
Tytera have produced several DMR portables, and the MD-380 has gained a good reputation. I borrowed one for a trial, and it performed quite well… well enough to proceed with purchase of an MD-390 which appears to have similar internals but revised packaging to obtain IP67 protection.
The MD-390 was purchased on eBay for $162 delivered. It came with a US power pack (earning bad feedback), programming cable, two antennas, an earpiece mic set, a disk of all sorts of drivers which aren’t needed, out of date CPS software and Chenglish manual. Continue reading Tytera MD-390 DMR portable evaluation
I bought a remote speaker-microphone (RSM) for a MD-390 DMR portable from 409shop.com, a 41-80K.
They assured me it was compatible with the radio in digital mode, but it turned out to be lousy with ‘motorboat noise’ on tx audio due to RF ingress tot he electret capsule.
Since the RSM was otherwise a good rugged and economical product, it was worth trying to rectify the RF ingress problem.
Above is a pic of the electret. Two fine tracks can be seen bonding the metal can of the electret to the -ve pin, so that is good… the can showed low resistance to the -ve pin. The +ve line is bypassed to the -ve line about 12mm from the electret with an unknown capacitor, but it was clearly not effective at 440MHz. Continue reading Another RFI mod of a speaker mic (41-80K) for DMR use
In a recent long running thread on impedance matching on one of the online fora, one poster offered the Ten-tec 540 manual as a reference for clarity on the subject (which of course got murkier with every posting as contributors added their version to the discussion).
The Ten-tec 540 was made in the late 1970s, one of the early radios with a solid state PA, and their manual give the
Technical facts of life to guide new owners to successful exploitation of this new technology.
technical facts of life is this little gem:
The standing wave ratio is a direct measure of the ratio between two impedances, ie an SWR of 3 to 1 tells us that one impedance is three times the other. Therefore the unknown impedance can be three times as large or three times as small as the known one. If the desired impedance that the transceiver wants to see is 50 ohms, and SWR of 3 to 1 on the line may mean a load impedance of either 150 or 17 ohms. …
This says that the SWR wrt 50Ω implies just two possible impedances, he is very wrong… it implies an infinite set of possible impedances. Continue reading Ten-tec on the meaning of SWR
Remote speaker-microphones and DMR portables discussed RF ingress to Speaker Mics(RSM) used with DMR radios in digital mode.
I purchased a Retevis SM that was advertised as original equipment for the MD-380, but turned out to be lousy with RF interference in the form of the ‘motorboat noise’ on transmit audio.
Dismantling the SM (fighting the way through TORX with PIN screws, what are they trying to protect) I found there is precious little RF filtering, just a single SMD cap at the end of a long (wrt 500MHz) branch track.
Above is the modified SM. Continue reading Retevis MD-380 speaker mic modification to reduce RF interference