Measure MLL using the half ReturnLoss method – a spot test with a hand held analyser
At Exploiting your antenna analyser #14 I gave an explanation of the method of approximating MLL of a line section by taking the average half Return Loss with o/c and s/c terminations.
This article demonstrates the technique using the Rigexpert AA-600 analyser in hand held mode.
The task is to assess whether a section of RG58A/U coax has MLL at 3.5MHz similar to specification or not.
The specification loss of 10.13m of RG58A/U has MLL=0.29dB.
Above, the first test with an o/c termination. Return Loss is 0.4dB. Continue reading Exploiting your antenna analyser #15
Assessing the Q of a half wave dipole antenna system explained that Q can be a valuable indicator of antenna system health.
This article uses a recently published VSWR curve for a 15m half wave dipole antenna system as an example to demonstrate the technique.
The following graph is from a Sark100 style antenna analyser, and it is quite a poor start to diagnostics, but using it draws out what is desired for further analysis.
Above, the captured VSWR(50) sweep. Continue reading Assessing the Q of a half wave dipole antenna system – a real world example
Q can be a valuable indicator of antenna system health
The Q of an antenna can be a useful statistic in assessing whether it is operating as it should.
The Q of half wave dipole antenna system on HF depends to some extent on conductor size, its environment (height, type of ground, nearby structures and vegetation, and feed line / matching loss. Nevertheless, it should usually fall in the range of 10 to 13 for good wire dipoles, and if you measure a half wave dipole antenna system to have Q significantly outside that range, it is probably significantly less efficient than it should be. Continue reading Assessing the Q of a half wave dipole antenna system
This article describes a build of the PIC Iambic Keyer (PIK).
Above is the generic circuit diagram of the PIK.
This one runs on 4.5V from 3 x AA cells. A 3000mAh battery will run it in ‘sleep’ mode for around 2,000,000 hours or 230 years… the shelf life of the batteries determines their useful life and there is consequently no ON/OFF switch.
So, the variation to the circuit above is that the zener regulator circuit is not required, Z1 is omitted and R5 is replaced by a 50mA Polyfuse. C3 is 0.0068µF to give a range of 6-36WPM on 4.5V.
Above, the internals. The electronics is assembled on a small piece of Veroboard with jacks at the rear for paddle, hand key and output, a pot for speed control and switches for TUNE and AutoSpace.
Above is the external view of the keyer prior to labelling.
At HC-500 I showed some VNA plots of the HC-500 matching a 50+j0Ω load at 3.5MHz.
The following commentary is on a single load scenario, a 50+j0Ω load at 3.5MHz, and while the results are not simply extensible to other loads and frequencies, it does provide some interesting insight into the devices.
THP HC-500 (Ultimate Transmatch (McCoy 1970))
Above is the behaviour of the unmodified HC-500 (an Ultimate Transmatch).
Losses at match are 12% of input power. At its rated 500W maximum power, that is 60W (which might seem high but heat tolerant insulation materials are used). On modification to a T match, losses at match were reduced to 8% or 40W at rated maximum power.
Continue reading A tale of three tuners
In the early 1970s I purchased a Tokyo High Power Labs HC-500 ATU based on recommendation of other hams and the seller’s representations (Dick Smith Electronics) that it was a T match with 200pF capacitors.
The circuit configuration is of the so-called Ultimate Transmatch, an invention of (McCoy 1970) that claimed a bunch of advantages over the ordinary T match.
The HC-2500 would appear to use the same circuit.
It wasn’t long before several authors waded into the Ultimate Transmatch over its poorer efficiency. With an ambitious name like Ultimate Transmatch, it had a lot to live up to… but it failed.
Within months, a reconfigured topology appeared entitles the SPC Transmatch, but it also had issues.
The reality is that none of these designs is ‘ultimate’, they all have advantages and disadvantages and are mostly used in ignorance of those.
So, I have had this HC-500 which worked well enough I suppose, but was quite difficult to tune on some loads that ordinary T matches handled with ease. It has always been my intention to reconfigure it to a T match be rewiring the grounded stator of the input cap to parallel it with the other stator… a minimal modification to get rid of the shunt capacitor and use it to help to keep coil voltage down on some loads.
Before performing the modification, I measured transmission loss when matched to a 50+j0Ω load at 3.5MHz using a two port VNA.
Above, transmission loss is 0.54dB, efficiency is 88.3%. Continue reading HC-500
At Improved cooling for the MFJ-949E I described a modification to the ATU to improve its cooling using a fan and run on timer.
The run on timer described was based on a Chinese STC15F104E DIP8 8051 like microcontroller.
Because the programming tools for the STC chips work so poorly, and the lack of documentation of their protocol, there is no simple way to update only the calibration data in EEPROM. I have ported the algorithm to an ATTINY25 which doesn’t cost a lot more but had a much better development environment and a range of tools to allow EEPROM update without overwriting the FLASH image, and as well it will run my bootloader, ATB.
This article describes a generic run on timer based on an Atmel AVR chip, a ATTINY25 though the code will also run in ATTINY45 and ATTINY85.
The circuit is very simple, the DC output from the forward power detector is connected to the input pin which turns the BC548C transistor on at input voltage greater than about 0.7V. The high value of base resistor ensures very light loading of the forward power detector.
Continue reading A generic run on timer using an ATTINY25
Sixth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
This article documents measurement of impedance.
The antenna system is a G5RV with tuned feeders (9m of home made 450Ω open wire). The tuned feeders terminate on the balun described in this series, and it is located on the outside of the antenna feed entrance panel shown above. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #6
Fifth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
Installation / testing
The balun packaged in a non-conductive housing was designed to have minimal stray capacitance to ground to minimise common mode current with asymmetric loads.
Above, the balun is attached to the exterior side of the antenna feed entrance panel using a male to male N adapter, done up very tight. The feed line connections are liberally coated with marine grease to prevent ingress of water and oxygen, a measure to reduce corrosion. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #5
At A look at internal losses in a typical ATU I demonstrated that it is quite easy to raise the temperature of the coil in the MFJ-949E to an unsafe level, even with quite modest power.
The most heat sensitive component in this ATU is the coil, specifically the coil supports which are probably polystyrene, and the melting temperature of polystyrene is around 100°.
This article documents modification of my MFJ-949E to reduce the risk of damage under some operating conditions. Continue reading Improved cooling for the MFJ-949E