## Critically review your measurements

A recent post on an online forum provides a relevant example to discussion of this subject.

I have personally seen ratios similar to 3:1 or higher at the feed point become 1:1 at the rig over 100 or so feet of coax cable.

First point is that in good transmission line, it takes an infinite length to deliver the observations made above. Less might deliver almost VSWR=1 at the input end of the line.

Let us consider a practical scenario, 100′ of RG58A/U with a load of 150+j0Ω at 14MHz, the load end VSWR(50) is 3, the input impedance is 32.50-j22.86Ω and input VSWR(50) is 2.01. In this scenario, the line loss is 2.5dB which might be unacceptable for some applications. Continue reading Exploiting your antenna analyser #19