The sign of reactance – challenge reality check

The sign of reactance – a challenge posed a problem, a set of R,|X| data taken with an analyser of a quite simple network and asked readers to solve the sign of X over the range, ie to transform R,|X| to  R,X.

The sign of reactance – challenge solution gave a solution to the challenge, and The sign of reactance – challenge discussion provided some discussion about the problem and solution.

Some correspondents have asserted that the challenge (see above Smith chart) contains a response that is contrived for the purpose and not representative of real world antenna systems. Continue reading The sign of reactance – challenge reality check

The sign of reactance – challenge discussion

The sign of reactance – a challenge posed a problem, a set of R,|X| data taken with an analyser of a quite simple network and asked readers to solve the sign of X over the range, ie to transform R,|X| to  R,X.

It is widely held that this is a trivial matter, and lots of software / firmware implement algorithms that fail on some scenarios. Though the scenario posed was designed to be a small set that provides a challenging problem, it is not purely theoretical, the characteristics of the data occur commonly in real world problems and the challenge data is derived from measurement of a real network.

Above is a Smith chart plot of the measured data that was transformed to the R,|X| for the challenge. Continue reading The sign of reactance – challenge discussion

The sign of reactance – challenge solution

The sign of reactance – a challenge posed a problem, a set of R,|X| data taken with an analyser of a quite simple network and asked readers to solve the sign of X over the range, ie to transform R,|X| to  R,X.

It is widely held that this is a trivial matter, and lots of software / firmware implement algorithms that fail on some scenarios. Though the scenario posed was designed to be a small set that provides a challenging problem, it is not purely theoretical, the characteristics of the data occur commonly in real world problems and the challenge data is derived from measurement of a real network.

Imported and rendered graphically in ZPlots we have:

The network measured is comprised from analyser, a 2.8m length of RG58/CU, a tee piece feeding a 50 resistor on one branch and on the other branch, another 2.8m length of RG58/CU with a 4.7Ω resistor termination.

The challenge is: what is the sign of X across the frequency range? Continue reading The sign of reactance – challenge solution

The sign of reactance – a challenge

Over time, readers of The sign of reactance have suggested that determining the sign of reactance with an antenna analyser that does not directly measure the sign is not all that difficult, even for beginners. The article shoots down some of the most common algorithms as failures on simple cases.

This article gives measurements made from a simple network of two identical lengths of 50Ω coax, a 50Ω resistor and a 4.7Ω resistor. It is a network designed to offer a challenge to the simple algorithms, and it IS solvable analytically… but not with most algorithms and software,

Here is the data from measurement made with an AA-600 and then all – signs removed, so in fact the Xs column is |Xs|.

```"Zplots file generated by AntScope"
"Freq(MHz)","Rs","Xs"
9.000000,78.13,53.66
9.250000,82.12,51.10
9.500000,86.10,47.83
9.750000,89.46,44.00
10.000000,92.30,39.90
10.250000,94.53,35.39
10.500000,96.21,30.71
10.750000,97.17,26.14
11.000000,97.49,21.54
11.250000,97.30,17.12
11.500000,96.54,13.04
11.750000,95.47,9.14
12.000000,93.92,5.68
12.250000,92.16,2.70
12.500000,90.25,0.17
12.750000,88.13,2.50
13.000000,85.94,4.50
13.250000,83.67,6.15
13.500000,81.45,7.36
13.750000,79.29,8.38
14.000000,77.22,9.21
14.250000,75.21,9.78
14.500000,73.23,10.16
14.750000,71.44,10.37
15.000000,69.70,10.25
15.250000,67.99,10.23
15.500000,66.50,9.99
15.750000,65.10,9.68
16.000000,63.81,9.27
16.250000,62.65,8.72
16.500000,61.59,8.15
16.750000,60.55,7.54
17.000000,59.69,6.86
17.250000,58.97,6.20
17.500000,58.20,5.43
17.750000,57.66,4.68
18.000000,57.14,3.81
18.250000,56.77,2.98
18.500000,56.47,2.16
18.750000,56.22,1.22
19.000000,56.04,0.38
19.250000,56.07,0.50
19.500000,56.02,1.38
19.750000,56.12,2.29
20.000000,56.41,3.15
20.250000,56.68,4.03
20.500000,57.11,4.86
20.750000,57.51,5.72
21.000000,58.06,6.61
21.250000,58.77,7.45
21.500000,59.54,8.22
21.750000,60.47,8.95
22.000000,61.44,9.75
22.250000,62.52,10.34
22.500000,63.77,10.97
22.750000,65.11,11.55
23.000000,66.56,12.02
23.250000,68.11,12.38
23.500000,69.82,12.64
23.750000,71.75,12.82
24.000000,73.67,12.84
24.250000,75.96,12.67
24.500000,78.12,12.27
24.750000,80.40,11.72
25.000000,83.05,10.69
25.250000,85.56,9.68
25.500000,88.29,8.09
25.750000,90.92,6.21
26.000000,93.63,3.91
26.250000,96.17,1.13
26.500000,98.61,2.16
26.750000,100.68,5.92
27.000000,102.51,10.11
27.250000,103.87,14.90
27.500000,104.65,19.98
27.750000,104.71,25.32
28.000000,103.98,30.95
28.250000,102.58,36.48
28.500000,100.14,41.97
28.750000,97.08,47.32
29.000000,93.07,51.86
```

Imported and rendered graphically in ZPlots we have:

The challenge is what is the sign of X across the frequency range? Continue reading The sign of reactance – a challenge

Ellington 3 x FT240-52 matching transformer for an EFHW

Ellington describes in a Youtube video his high power matching transformer for an EFHW, he rates it suited to 500W CW.

Like almost all such ‘designs’, they are published without supporting measurements or simulations.

The transformer is intended to be used with a load such that the input impedance Zin is approximately 50+j0Ω, Gin=0.02S.

Analysis of a simple model of the transformer with a load such that input impedance is 50+j0Ω gives insight into likely core losses.
Continue reading Ellington 3 x FT240-52 matching transformer for an EFHW

A flexible test panel for microcontroller based power control projects – #2

This article expands on A flexible test panel for microcontroller based power control projects with some enhancements and accessories.

A LED power meter that I had ordered finally arrived (slow boat from China syndrome).

Above, the upper rail contains a RCD, the power meter which displays Volts, Amps, and kW, or pf, hours, and kWh, a DIN mount terminal block for mains, and a 40A SSR on a heatsink. A clip on CT can be used for oscilloscope observation of mains current. Continue reading A flexible test panel for microcontroller based power control projects – #2

Australian amateur population trends 1998 – 2017

This is a 2017 update of an article written originally in October 2005, earlier editions published on VK1OD.net which is now offline.

Over recent years to 2002, the number of issued amateur licences was declining, the trend was about 2.8% pa decline over the five years to 2002.

This has concerned some people, who took the view that the decline was a harbinger of the impending demise of Amateur Radio. Continue reading Australian amateur population trends 1998 – 2017

KL7AJ on the Conjugate Match Theorem – analytical solution – Winsmith

KL7AJ on the Conjugate Match Theorem asked the question Should we have expected this outcome?

Let us solve a very similar problem analytically where measurement errors do not contribute to the outcome.

Taking the load impedance to be the same 10.1+j0.2Ω, and calculating for a T match similar to the MFJ-949E (assuming L=26µH, QL=200, and ideal capacitors) we can find a near perfect match.

The capacitors are 177.2 and 92.93pF for the match.

Now turning the network around by swapping the capacitors and changing the load to 50+j0Ω. Continue reading KL7AJ on the Conjugate Match Theorem – analytical solution – Winsmith

KL7AJ on the Conjugate Match Theorem

KL7AJ proposed a little test for his readers on QRZ:

One of the most useful (and sometimes astonishing) principles in radio is the Conjugate Match theorem. In the simplest terms, what this says is that the maximum power will be transferred between a source (like a transmitter) and a load (like an antenna), when the source impedance is the COMPLEX CONJUGATE of the load impedance (or vice versa).
Here’s a neat little experiment to prove the conjugate match theorem. You need four basic ingredients: an antenna analyzer like the MFJ259 (or an actual impedance bridge, if you know how to use one). A good low loss antenna tuner. A good 50 ohm resistor. And a good 200 ohm resistor. And some appropriate connecting hardware, namely some short bits of coax.

Step 1) connect the 50 ohm resistor to the OUTPUT of the antenna tuner. Connect the antenna analyzer to the INPUT of the antenna tuner.

Step 2) Adjust the antenna tuner to get precisely 50 ohms, zero reactance on the antenna analyzer. This step simply confirms everything is working.

Step 3) Replace the 50 ohm resistor with the 200 ohm resistor. Readjust the antenna tuner to get 50 ohms, zero reactance on the antenna analyzer. Do not disturb the antenna tuner adjustments after this point.

Step 4) Remove the 200 ohm resistor and insert the antenna analyzer in its place (at the OUTPUT of the antenna tuner).

Step 5) Insert the 50 ohm resistor at the INPUT of the antenna tuner.

Step 6) Take a careful reading of the antenna analyzer. (What do you think it will say?)

10 points for anyone who will correctly explain why this works.

Some clarifications

Jacobi maximum power transfer theorem

Jacobi published his maximum power transfer theorem in 1840. It states that maximum power is transferred from a (Thevenin) source to a load when the load resistance is equal to the (Thevenin equivalent) source resistance.

It was later adapted to apply to AC circuits with sinusoidal excitation, maximum power is transferred from a (Thevenin) source to a load when the load impedance is the complex conjugate of the (Thevenin equivalent) source impedance.

Walt Maxwell’s Conjugate Mirror

(Maxwell 2001 24.5) states

To expand on this definition, conjugate match means that if in one direction from a junction the impedance has the dimensions R + jX, then in the opposite direction the impedance will have the dimensions R − jX. Further paraphrasing of the theorem, when a conjugate match is accomplished at any of the junctions in the system, any reactance appearing at any junction is canceled by an equal and opposite reactance, which also includes any reactance appearing in the load, such as a non-resonant antenna. This reactance cancellation results in a net system reactance of zero, establishing resonance in the entire system. In this resonant condition the source delivers its maximum available power to the load. …(1)

Note that it states that if a conjugate match is established an any junction, then a conjugate match occurs in any (all) other junctions, simultaneously a conjugate match exists everywhere. Continue reading KL7AJ on the Conjugate Match Theorem