I repaced the R134a refrigerant in my car aircon system with a hydrocarbon refrigerant, Hychill Minus 30 (HC-30).
Research had indicated that permeation of reduced barrier hoses was problem with HC refrigerants. The hoses in the car were Goodyear hoses with catalogue numbers, but I was unable to find data on them. Their diameter was comparable to standard barrier hoses, so I proceeded with the trial. Continue reading Post post implementation review R134a replaced with HyChill Minus 30
The Ferrite permeability interpolations calculator performs interpolations of tables of complex permeability data.
From manufacturer’s curves
Some of the data is derived from manufacturer’s published complex permeability curves. The plot above shows the Ferroxcube’s published curve for 3C81 material, and points at which it was digitised to extract a table of µ’ and µ”. Continue reading Online calculator of ferrite material permeability interpolations – more detail
I have had a couple of Holux GPS loggers for some time, a RCV3000 and GPS245+.
They come with a neat little application called Holux ezTour for Logger.
Over time, the application has been unreliable, mostly knock on effects of Google Maps licencing and Holux’s failure to obtain and distribute the appropriate licence keys. Continue reading Holux web site missing / offline, Holux ezTour crippled
OpenLog for TinyTrak described a data logger for APRS NMEA data.
This article shows an innovative inline adapter for adding an Openlog to an existing APRS installation where the GPS is powered with 5V via the DE9 connector.
Above, the adapter comprises a DE9 male and DE9 female connector cabled together as a through connection, and the Openlog is wired into one of the backshells, wrapped in heat shrink tube, and velcro’d to the backshell. Continue reading OpenLog for TinyTrak – inline DE9 adapter
Recently I have had difficult reaching the local DMR repeater on 70cm, and needed to check that the antenna system had not deteriorated.
I took a baseline measurement with an AA-600 after some refurbishment work in Jan 2018, and was able to compare a current sweep to that baseline.
Above, a wide Return Loss sweep of the Diamond X-50N with feed line compared to the baseline (the thin blue line). Continue reading Diagnosing a possible antenna problem by comparison with a baseline
In my experience, action cam underwater / weatherproof transparent cases have buttons that are quite difficult to use. This seems mostly because they are assembled with no lubricant.
Above, the button in the genuine Sjcam SJ8 has a stainless spring, shaped washer, two 6x2x2mm orings, and a 2.5mm E clip. This pic shows that the thing is assembled dry which makes operating pressure high, and in any event not smooth as the button shaft is not particularly smooth. Continue reading Refurbishing the push button in action cam cases
A chap seeking details for a matching inductor for his 5/8λ vertical on 20m reported “my AA54 RigExpert analyser gives the following reading (SWR 8,2). (R 81,5). (X -158) ” measured looking into a “length of rg58 about 15-20 cm” and asked “is the inductor coil going to be enough or will I need an L match to bring the real resistance to 50 ohms”. Continue reading Matching a 5/8λ ground plane – a single stub tuner example
Some time ago I wrote some articles on so-called Coax Traps, and an example design of an Inverted V dipole for 80 and 40m.
A coax trap (before cross connection).
The whole subject of trapped antennas elicits a lot of online discussion that is often more about semantics than understanding. Continue reading Trapped dipole
Designing with ferrite binocular cores can be frustrating as there are different formats in which data is provided, and data for different mixes on the same dimensioned cores appear inconsistent.
There are several source of the Al parameter for some common cores, often from resellers rather than manufacturers. Continue reading Designing with binocular ferrite cores – published Al values