## Finding resistance and reactance with some low end analysers #2

Exploiting your antenna analyser #8 was about finding resistance and reactance with some low end analysers that don’t directly display those values of interest. The article showed how to calculate the values starting with |Z| from the analyser and included links to a calculator to perform the calcs.

This article describes an extension to that calculator Find |Z|,R,|X| from VSWR,|Z|,R,Ro to use R, VSWR, and Ro as the starting point. Note that the sign of X and the sign of the phase of Z cannot be determined from this starting point, there just isn’t enough information.

You will probably not find the equation for |X|(R,VSWR,Ro) in text books or handbooks, and the derivation is not shown here but if there is interest, I may publish a separate paper.

Lets say you knew VSWR=2, R=75Ω, Ro=50Ω, what is |X|?

Above, entering the values in the calculator we find that |X|=35.4Ω. Continue reading Exploiting your antenna analyser #20

## Analysis of a certain dipole animation – OCF implications

A correspondent having read Analysis of a certain dipole animation questioned the validity of the lossy transmission line model of the dipole, citing the case of an OCF half wave which has an approximately resistive feed point.

Since the OCF lacks the symmetry exploited in earlier study, we must consider each half of the OCF dipole and combine them. To assist, I have produced a similar plot of the transmission line but note the changed X axis.

The scenario is again a 2mm diameter copper wire, 3m above ground at 1MHz.

Z0 can be approximated as 138*log(2h/r)=138*log(2×3/0.001)=521Ω.

Above is a plot of calculated R,X components of V/I at displacements from the open end, and calculated phase of V/I. Continue reading Analysis of a certain dipole animation – OCF implications

## Exploiting your antenna analyser #19

A recent post on an online forum provides a relevant example to discussion of this subject.

I have personally seen ratios similar to 3:1 or higher at the feed point become 1:1 at the rig over 100 or so feet of coax cable.

First point is that in good transmission line, it takes an infinite length to deliver the observations made above. Less might deliver almost VSWR=1 at the input end of the line.

Let us consider a practical scenario, 100′ of RG58A/U with a load of 150+j0Ω at 14MHz, the load end VSWR50 is 3, the input impedance is 32.50-j22.86Ω and input VSWR50 is 2.01. In this scenario, the line loss is 2.5dB which might be unacceptable for some applications. Continue reading Exploiting your antenna analyser #19

## Analysis of a certain dipole animation

Modern people look for videos and animations for their learning, and these are often not from reputable sources and raise more questions than they answer.

An example is an animation of a half wave dipole on the Internet, and being discussed on QRZ.

Above, the animated graphic.

Without trying to understand the problem, lets just extract two cases for further discussion, an analysis in the limits if you like. Continue reading Analysis of a certain dipole animation

## Measure velocity factor of open wire line

One of the measurement tasks that one often encounters is to measure the velocity factor of a transmission line.

Often this is an indirect task of tuning a tuned line section, my method is to often measure some line off the role, find the velocity factor (vf), and use that to cut line for the tuned section making appropriate allowance for connectors etc.

Measuring vf for an open wire line includes all that is done for measuring vf of coax, but requires measures to ensure that common mode current does not affect measurement significantly.

To minimise common mode current effects, I will use two measures:

• a high common mode impedance Guanella balun; and
• form the line section being measured into a loose helix supported on some fishing line to spoil any common mode resonance.

Above is the balun used, it is described at Low power Guanella 1:1 balun with low Insertion VSWR using a pair of Jaycar LF1260 suppression sleeves. Continue reading Exploiting your antenna analyser #18

## Optimising a G5RV with hybrid feed

(Varney 1958) described his G5RV antenna in two forms, one with tuned feeders, and the more popular form with hybrid feed consisting of a so-called matching section of open wire line and then an arbitrary length of lower Z0 coax or twin to the transmitter.

(Duffy 2005) showed that the hybrid feed configuration is susceptible to high losses in the low Z0 line as it is often longish, is relatively high loss line and operates with standing waves. Varney did offer two options for the low Z0 line: any length 72Ω twin or coax. Continue reading Exploiting your antenna analyser #17

## Measure inductor using SOL calibration

At Measuring balun common mode impedance I showed a method of backing out the effects of a text fixture using the “subtract cable” facility of Antscope software with the Rigexpert AA-600.

Some analysers (including the AA-600) support SOL calibration of the instrument itself, and some support SOL calibration using the client software (Antscope in this case). This article demonstrates use of Antscope with SOL calibration to measure a small RF inductor which has similar characteristic to good Guanella 1:1 HF baluns.

The text fixture used for this demonstration is constructed on a SMA(F) PCB connector using some machined pin connector strip, and SMA(M)-SMA(M) and N(M)-SMA(F) adapters to connect to the AA-600.

Above is a pic of the test inductor in the test fixture with adapters. The test inductor 6 turns of 0.5mm PVC insulated wire wound on a BN43-202 binocular balun core. Continue reading Exploiting your antenna analyser #16

## Improved cooling for the ATR-30

In Improved cooling for the MFJ-949E I described a solution to a problem of demonstrated overheating of the ATU at rated power, indeed at a lot less than rated power.

Though I have never measured the ATR-30 temperature rise, and am probably unlikely to stress the 3kW rated ATU with a 100W transmitter, I have performed a similar cooling modification to the ATR-30.

## Measure MLL using the half ReturnLoss method – a spot test with a hand held analyser

At Exploiting your antenna analyser #14 I gave an explanation of the method of approximating MLL of a line section by taking the average half Return Loss with o/c and s/c terminations.

This article demonstrates the technique using the Rigexpert AA-600 analyser in hand held mode.

The task is to assess whether a section of RG58A/U coax has MLL at 3.5MHz similar to specification or not.

The specification loss of 10.13m of RG58A/U has MLL=0.29dB.

Above, the first test with an o/c termination. Return Loss is 0.4dB. Continue reading Exploiting your antenna analyser #15

## A generic run on timer using an ATTINY25

At Improved cooling for the MFJ-949E I described a modification to the ATU to improve its cooling using a fan and run on timer.

The run on timer described was based on a Chinese STC15F104E DIP8 8051 like microcontroller.

Because the programming tools for the STC chips work so poorly, and the lack of documentation of their protocol, there is no simple way to update only the calibration data in EEPROM. I have ported the algorithm to an ATTINY25 which doesn’t cost a lot more but had a much better development environment and a range of tools to allow EEPROM update without overwriting the FLASH image, and as well it will run my bootloader, ATB.

This article describes a generic run on timer based on an Atmel AVR chip, a ATTINY25 though the code will also run in ATTINY45 and ATTINY85.

The circuit is very simple, the DC output from the forward power detector is connected to the input pin which turns the BC548C transistor on at input voltage greater than about 0.7V. The high value of base resistor ensures very light loading of the forward power detector.
Continue reading A generic run on timer using an ATTINY25