K0KA made a quite polished presentation published as a Youtube video explaining how to measure balance performance of an antenna system and an ATU.
He did not define what he means by balance… but it will become apparent.
Midway through his video, he measures the ‘balance’ of a Palstar AT5K (though possibly modified) by connecting a symmetric load consisting of two 470Ω 2% resistors in series and connecting the junction to the ground terminal on the ATU.
Above, from KA0KA’s video, his test load. The oscilloscope probes can be seen.
The oscilloscope channel gains are carefully adjusted to be equal.
VU3SQM offers an interesting directional coupler based on a Sontheimer coupler, and using AD8307 power sensing for a nominally HF coupler. I must say that I am not a fan of Sontheimer couplers… but that is what the board uses.
This article lays out a preliminary design review to assist in selection of appropriate toroids, and ordering of the needed parts.
Standard Plus Evening Speed—plans using this label will deliver a minimum speed of 30Mbps during the busy period. This plan would be suitable for a higher usage profile (e.g. streaming an ultra-high definition movie and streaming music on one or more other device during the busy period)
Before trusting measurements made with any instrument, its behaviour should be validated, and this article documents issues discovered in one thread of tests. The developer does not like the term “defects”, he prefers “issues”, a soft denial of “problems”.
So, the test scenario is the VA5 measuring the impedance looking into a 35m length of RG6 coax with an open circuit at the far end. The VA5 has been SOL calibrated with the higher quality loads sold by SDR-kits, and the test is a 3.5MHz. The firmware is the latest, v1.08 (about 3 months old).
The screenshots are taken with a camera, there does not seem to be a method of uploading screenshots to a PC.
I have an inexpensive (~$10 inc post) digital temperature controller that was observed to have very high RF emissions and was unusable because it was incompatible with radio operations here.
There has been enhanced firmware for a STC-1000 controller based on a PIC chip, but in my experience, most seem to be made with an STM8S003F3 chip. More recently, what appears to be a port of the PIC software to STM8 has been published at https://github.com/Emile666/stc1000_stm8.
The enhanced firmware is directed at home brewing for control of long running fermentation processes etc, incorporating storage for a number of multi-step programs (profiles). It is not really suited to more general applications like a fridge controller as for example it is more difficult for the common operation to set the set point.
I thought it might be entertaining to try it out.
Installation of the firmware was straight forward, I had the necessary programming adapter and software.
Interestingly, after updating the firmware, the device would not start on its original power supply. My suspicion is that the new firmware runs the LED display at higher duty cycle and the power supply could not withstand the load.
Since the power supply had RF emission problems, it was removed and the device converted to 12V DC.
Above, the large components of the switched mode power supply were removed, a Polyswitch protective device installed and input routed to the output terminals of the old 12V output switched mode power supply (white wire is -ve, brown wire is +ve). Continue reading STC-1000 firmware change
On Thevenin’s theorem looked at a simple source network to demonstrate some key characteristics and limitations of Thevenin’s equivalent circuit.
The example network used was linear in V,I for all V,I combinations possible. Let’s now look at a network that is not linear for all V,I, but is sufficiently linear over a sub range to be usefully modelled using Thevenin’s equivalent circuit.
Black Box for discussion
For the purpose of discussion, we have a Black Box with just two terminals and is a source of DC voltage and current, and the internal implementation is hidden from us.
A series of measurements is made with different load resistors attached and the voltage and current at the terminals is recorded and plotted uniformly stepped currents.
The V,I characteristic is clearly non-linear, but on closer examination there are two fairly linear regions, from 0.008 to 0.060A and 0.08A to 0.1A. It is a device that is usually used in the region below the knee, and for our application, let us concentrate on 0.008 to 0.030A. Continue reading On Thevenin’s theorem – #2
Whilst preparing A first test of the FA-VA5 antenna analyser, issues were noticed with the user interface design / implementation. I stated in a later article that The matter of the clumsy / unproductive user interface will be explored more at a later time.
This article introduces a short video demonstration of the frustrating / unreliable user interface (UI) in firmware v1.08 where buttons do not seem to operate intuitively and consistently.