There are many ways to get a good estimate of the characteristic impedance Zo of a transmission line.

One method is to measure the input impedances of a section of line with both a short circuit and open circuit termination. From Zsc and Zoc we can calculate the Zo, and the complex propagation constant \(\gamma=\alpha + \jmath \beta\).

Calculation of Zo is quite straightforward.

The solution for γ involves the log of a complex number \(r \angle \theta\) which is one of the many possible values \(ln(r) + j \left(\theta + 2 \pi k \right)\) for +ve integer k. Conveniently, the real part α is simply \(ln(r) \). The real part of γ is the attenuation in Np/m which can be scaled to dB/m, and the imaginary part is the phase velocity in ** ^{c}**/m. The challenge is finding k.

## Measurement with nanoVNA

So, let’s measure a sample of 14×0.14, 0.22mm^2, 0.5mm dia PVC insulated small speaker twin.

Above is the nanoVNA setup for measurement. Continue reading Measure transmission line Zo – nanoVNA – PVC speaker twin