The findings at InsertionVSWR of Revex W560 on HF and the suggestion that the low frequency problem is characteristic of poorly designed Sontheimer couplers (Sontheimer, C & Frederick 1966).
These couplers were popularised by (Grebenkemper 1987) in his Tandem Match – An Accurate Directional Wattmeter and have appeared in ARRL handbooks over the decades, and may have inspired the many commercial implementations of the coupler.
Grebenkemper claims his meter is ‘good’ down to 1.8MHz, but does not clearly claim any particular InsertionVSWR. There is limited value in an instrument that can measure down to 1.05 when it causes significantly higher VSWR itself.
Lets drill down on Gebenkember’s article, specifically the coupler design.
Continue reading InsertionVSWR of Grebenkemper’s Tandem Match
The Revex W560 is a dual range VSWR meter that was also sold under other brand names.
The low frequency range is specified as 1.8-160MHz. Continue reading InsertionVSWR of Revex W560 on HF
Optimal Zo for Guanella balun sections left the reader with a though exercise, a transmission line transformer used by PA0V in a 144MHz power amplifier output network.
The pair of tabs to the left are driven by FET drains, the upper pink centre conductor is grounded, the lower end connecting to C1 is the output to a nominal 50R load. The network shown near OUT is for fine load adjustment. There are two coax sections making this TLT, shields bonded all the way around and the centre conductors connected as shown. What is the optimal value of Zo for each the coax sections?
Above is a pic of the PA, and we are looking at the network to the right of the dual FET. Continue reading Optimal Zo for TLT sections challenge – a solution
A Guanella balun may have several sections, and they may be connected in parallel on one side and series on the other side so as to achieve nominal impedance transformation ratios other than 1.
The question is often asked, what is the optimal Zo for these line sections?
Several answers exist in ham lore, but the answer is relatively simple and revealed by the most basic understanding of transmission lines.
If you do not want standing waves on a line section and its associated impedance transformation, then make sure that Zo=V/I… easy as that.
(Guanella 1944) explains it with examples:
Note above that he refers to
coil systems. He did not describe for instance (b) on a single core, a shared magnetic circuit which would be a single core system, but he states clearly
two coil systems. (Sevick 2001) and lots of other hams say otherwise, but they are wrong. Continue reading Optimal Zo for Guanella balun sections
Some recent articles here used a two port analyser to evaluate Insertion VSWR of some coax switches, and it raises the question about application of a hand held analyser and Insertion VSWR of a VSWR meter.
(Duffy 2007) listed tests for evaluation of a VSWR meter:
Testing a VSWR meter
The tests here need to be interpreted in the context of whether the device under test (DUT) has only calibrated power scales, or a VSWR Set/Reflected mode of measurement, and whether directional coupler scales are identical for both directions.
- Connect a calibrated dummy load of the nominal impedance on the instrument output and measure the VSWR at upper and lower limit frequencies and some in between frequencies. The VSWR should be 1. (Checks nominal calibration impedance);
- Repeat Test 1 at a selection of test frequencies and for each test, without changing transmitter power, reverse the DUT and verify that repeat the forward/set and reflected readings swap, but are of the same amplitude (checks the symmetry / balance of the detectors under matched line conditions).
- Connect a s/c to the instrument output and measure the VSWR at upper and lower limit frequencies and some in between frequencies. The VSWR should be infinite. (Discloses averaging due to excessive sampler length);
- Connect an o/c to the instrument output and measure the VSWR at upper and lower limit frequencies and some in between frequencies. The VSWR should be infinite. (Discloses averaging due to excessive sampler length);
- Connect a calibrated wattmeter / dummy load of the nominal impedance on the instrument output and measure calibration accuracy of power / ρ / VSWR scales at a range of power levels in both forward and reflected directions (Checks scale shape and absolute power calibration accuracy).
- Repeating Test 1 additionally with a calibrated VSWR meter connected to the input to the DUT, and measure the VSWR caused by the DUT at a range of test frequencies (Checks Insertion VSWR).
It is not unusual for low grade instruments to pass Test 1, but to fail Test 6 (and some others, especially Test 3 and Test 4) towards the higher end of their specified frequency range.
Item 6 in the list was to evaluate the Insertion VSWR. Continue reading Can a hand held analyser be used to evaluate Insertion VSWR of a VSWR meter?
At Ratings of coax antenna switches I showed characteristics of a home made switch which has very low InsertionVSWR, but poor isolation.
A couple of correspondents have offered an explanation that the unused port must be shorted to get good isolation.
If that was the case, then we would expect all coax switches that leave the unused port open to have poor isolation.
Let us look at a very good coaxial relay
Above is a Dowkey 402 series relay which has good performance to GHz. It does not short the unused port.
Continue reading Coax switches – is shorting unused port necessary for isolation?
In a recent article I discussed how InsertionLoss implies InsertionVSWR in lossless devices.
This article looks at measurements of a few antenna switches at hand.
Daiwa CS-201G II
It is difficult to find comprehensive data on the very popular Daiwa CS-201 series switches.
Above is the data from the packet of one of these switches, a CS-201G II. The specifications are pretty loose, and one must depend on one’s own measurements.
Above, the CS-201G II, a basic CS-201 series switch with N connectors, advertised as useful to 2000MHz where InsertionLoss is given as 1.2dB (or better?). If there were no TransmissionLoss in the switch, that would imply InsertionVSWR=3.6, but there is probably some significant TransmissionLoss and InsertionVSWR would be somewhat less. Nevertheless, IMHO InsertionLoss=1.2dB indicates it as unsuitable such frequencies. Continue reading Ratings of coax antenna switches
Devices inserted in transmission lines often characterised by one or more of:
- Insertion VSWR (the input VSWR when terminated with a matched load);
- Return Loss (RL) in dB (20 times the log of the magnitude of the complex reflection coefficient); and
- Insertion Loss.
Practitioners often find Insertion VSWR (1) of most use as it indicates whether the device is worse than other system devices, the weak link in the chain if you like. You might see a coax antenna switch specified to have InsertionVSWR<1.2 to 60MHz.
Return Loss (2) is a function of VSWR and vice versa, so it appeals when the designer thinks in terms of Return Loss rather than VSWR (and it is a better metric for VSWR<1.2). You might see a coaxial relay specified to have ReturnLoss>30dB to 500MHz.
Insertion Loss (3) is not so readily compared to the other two which are measures of input reflection with a matched termination. It often yields some numbers that appear very acceptable, but might be deceptively so. You might see a coaxial relay specified to have ReturnLoss>30dB to 500MHz. You might see a coax antenna switch specified to have InsertionLoss<0.2dB to 100MHz. Continue reading InsertionLoss implies InsertionVSWR in lossless devices
On a transmission line with standing waves, the voltage varies cyclically along the line, and is dependent also on power.
This article explains a method to use an analyser to predict the peak voltage level at a point for a given frequency and power based on measurement or estimation of complex Z or Y at that point using a suitable antenna analyser.
Lets say you have some critical voltage breakdown limit and want to use your analyser to find any non-compliance at the proposed power level.
Let us assume that the not-to-exceed voltage at that point is 1000Vpk. Let’s allow a little margin for variation due to factors not fixed, let’s actually use 800Vpk as the limit. We will use the maximum permitted power in Australia, 400W.
Continue reading Exploiting your antenna analyser #22
The popular End Fed Half Wave is all things to all men, but this article compares an End Fed Half Wave, Inverted L, and Half Wave Dipole with some common parameters:
- frequency: 7.1MHz;
- flat top length: 20m;
- Height above ‘average’ ground (σ=0.005, εr=13): 10m;
- lossless balun / matching device.
- ground connection: Inverted L = 2Ω, End Fed Half Wave = 100Ω; and
- effective common mode choke used on the dipole.
Above is the modelled gain for all three. Continue reading End Fed Half Wave / Inverted L / Half Wave Dipole