## Transformer loss from half return loss

The popularity of the nanovna with new users has led to a lot of ‘beginners’ seeking advice of the online experts.

A recent question and the responses exposes some common misunderstanding / woolly thinking.

A beginner in search of wisdom asked if there an easy way to measure balun loss?

One of the responses was…

Measure the return loss of the balun with the balun shorted. The return loss should be about twice the balun loss. This is similar to measuring the loss of a shorted or open piece of cable.

This was expanded by others, basically supporting the concept. Continue reading Transformer loss from half return loss

## nanoVNA-H – another hardware fail – battery socket

I have a nanoVNA-H which has had many hardware problems, some designed in,but mostly sub-standard / faulty components.

Above, the latest repair. A new battery socket to replace the original that crumbled apart… sub-standard plastic from all appearances. This was from a reputable supplier, so it is probably a genuine Molex Picoblade part rather than some cheap Chinese knock off.

The blue wire is part of a mod to invoke the bootloader on power up, R5 was also changed to something small, 1k IIRC.

PS: a word of warning… always check polarity when fitting a battery, there is not rigid standardisation of connectors on LIPO batteries.

## SND implications of variations on the untuned small loop at MF/lowHF

Three recent articles developed an explanation of the YouLoop-2T at MF/lowHF:

The first and third articles explained the concept of signal/noise degradation (SND) statistic, and gave graphs of the behavior of the subject antennas.

This article draws together those SND plots for two antennas, and some variations to the configurations.

## Configurations

### Simple loop with transformer

Above, the “simple loop” with 0.5:1 ideal transformer. It could be implemented as a shielded loop (with transformer) with similar behavior (but improved common mode suppression). Continue reading SND implications of variations on the untuned small loop at MF/lowHF

## Osram Substitube tear down

I recently purchased two packs of 2x Osram Substitube LED replacements for a T8 36W florescent tube. The price per tube was about double that of a T8 fluorescent, and claimed life was 30,000 hours.

Note that LED life is usually an estimate of the time for 50% failure. Curiously, Osram individual LEDs have lower lifetime estimates for single LEDs, strings of LEDs will have lower lifetime, and lifetime for say 5% failures would be even lower. On the basis of experience with LED lighting, it might be optimistic to think that most of these lamps will last at least 5,000h.

In the event, three of four tubes had broken glass (yes, they use a glass tube much like the T8 fluorescent tube… though not sealed at the ends), and the other was DOA, no light output.

The LED driver is attached to pins at one end of the lamp, and covered by a label over the glass. Note that there are several incompatible schemes used in T8 fluorescent replacement LEDs, the other scheme bonds the adjacent pins at each end of the lamp and connects line and neutral to opposite ends. Osram calls this a “Type A” connection, and as the name suggests if can be fitted to an ordinary magnetic ballast luminaire PROVIDED the original starter is replaced with Osram’s “LED starter” which is actually a HRC fuse of around 1A rating.

Above is the top view of the electronics. Continue reading Osram Substitube tear down

## Towards understanding the YouLoop-2T at MF/lowHF

Small untuned loop for receiving – simple model with transformer gave a simple model for analysing a loop. If you haven’t already read it, you should. It provides a step towards understanding the YouLoop-2T at frequencies where is is a small loop (perimeter<λ/10).

Above is the Airspy Youloup-2T. Try to put the two turns thing out of your mind, it is misleading, panders to some common misunderstanding, and so does not help understanding.

This is somewhat similar to the simple loop, but now the transformer primary is connected to the loop gap terminals by two parallel sections of 50Ω transmission line, the combination being effectively a 100Ω with similar parameters to the component coax sections. Because of the series connection at the transformer and parallel connection at the loop gap, there is a 1:4 impedance transformation additional to that of the coax sections themselves. Continue reading Towards understanding the YouLoop-2T at MF/lowHF

## A transmission line 1:4 impedance transformer

This article explains the operation of a simple nominally 1:4 impedance transformer using transmission line (TL) elements.

Above is a diagram of the device. The currents shown are differential currents in the coax (ie wholly inside the coax), the current on the outside of the shield is not shown on the diagram.

At very low frequencies it may be intuitive that $$V_1\approx V_2$$ and $$I_1\approx I_2$$, but as frequency increases, a more exact solution is needed. Continue reading A transmission line 1:4 impedance transformer

## Small untuned loop for receiving – simple model with transformer

I have written several articles on untuned loops for receiving, as have others. A diversity of opinions abounds over several aspects, probably none more than the idea of an optimal load impedance for the loop.

This article analyses a simple untuned / unmatched loop in the context of a linear receive system (ie no IMD) of known Noise Figure. Continue reading Small untuned loop for receiving – simple model with transformer

## nanoVNA-H – Port 1 attenuator for improved what???

At nanoVNA-H – Port 2 attenuator for improved Return Loss I explained the reasons for essentially permanent attachment of a 10dB attenuator to Port 2 (Ch 1 in nanoVNA speak).

Above, the 10dB attenuator is semi permanently attached to Port 2 principally to improve the Return Loss (or impedance match) of Port 2, a parameter that becomes quite important when testing some types of networks than depend on proper termination (eg many filters). I should remind readers that the improvement in Port 2 Return Loss comes at a cost, the dynamic range of Port 2 is reduced by 10dB. Continue reading nanoVNA-H – Port 1 attenuator for improved what???

## Return Loss Bridge – some important details

Articles describing how to make a Return Loss Bridge are pretty common, but they don’t often spell out component values that are critical to accuracy.

Above is a schematic for discussion. It is somewhat simplified, but it is complete and will work. Continue reading Return Loss Bridge – some important details