## nanoVNA-H – measure ferrite transformer – Noelec balun

At nanoVNA-H – measure ferrite transformer I gave an example of using a nanovna to measure loss of a ferrite cored transformer.

Noelec makes a small transformer, the Balun One Nine, pictured above and they offer a set of |s11| and |s12| curves in a back to back test. (Note: back to back tests are not a very reliable test.) Continue reading nanoVNA-H – measure ferrite transformer – Noelec balun

## nanoVNA-H – another hardware fail – battery socket

I have a nanoVNA-H which has had many hardware problems, some designed in,but mostly sub-standard / faulty components.

Above, the latest repair. A new battery socket to replace the original that crumbled apart… sub-standard plastic from all appearances. This was from a reputable supplier, so it is probably a genuine Molex Picoblade part rather than some cheap Chinese knock off.

The blue wire is part of a mod to invoke the bootloader on power up, R5 was also changed to something small, 1k IIRC.

PS: a word of warning… always check polarity when fitting a battery, there is not rigid standardisation of connectors on LIPO batteries.

## SND implications of variations on the untuned small loop at MF/lowHF

Three recent articles developed an explanation of the YouLoop-2T at MF/lowHF:

The first and third articles explained the concept of signal/noise degradation (SND) statistic, and gave graphs of the behavior of the subject antennas.

This article draws together those SND plots for two antennas, and some variations to the configurations.

## Configurations

### Simple loop with transformer

Above, the “simple loop” with 0.5:1 ideal transformer. It could be implemented as a shielded loop (with transformer) with similar behavior (but improved common mode suppression). Continue reading SND implications of variations on the untuned small loop at MF/lowHF

## Towards understanding the YouLoop-2T at MF/lowHF

Small untuned loop for receiving – simple model with transformer gave a simple model for analysing a loop. If you haven’t already read it, you should. It provides a step towards understanding the YouLoop-2T at frequencies where is is a small loop (perimeter<λ/10).

Above is the Airspy Youloup-2T. Try to put the two turns thing out of your mind, it is misleading, panders to some common misunderstanding, and so does not help understanding.

This is somewhat similar to the simple loop, but now the transformer primary is connected to the loop gap terminals by two parallel sections of 50Ω transmission line, the combination being effectively a 100Ω with similar parameters to the component coax sections. Because of the series connection at the transformer and parallel connection at the loop gap, there is a 1:4 impedance transformation additional to that of the coax sections themselves. Continue reading Towards understanding the YouLoop-2T at MF/lowHF

## A transmission line 1:4 impedance transformer

This article explains the operation of a simple nominally 1:4 impedance transformer using transmission line (TL) elements.

Above is a diagram of the device. The currents shown are differential currents in the coax (ie wholly inside the coax), the current on the outside of the shield is not shown on the diagram.

At very low frequencies it may be intuitive that $$V_1\approx V_2$$ and $$I_1\approx I_2$$, but as frequency increases, a more exact solution is needed. Continue reading A transmission line 1:4 impedance transformer

## Small untuned loop for receiving – simple model with transformer

I have written several articles on untuned loops for receiving, as have others. A diversity of opinions abounds over several aspects, probably none more than the idea of an optimal load impedance for the loop.

This article analyses a simple untuned / unmatched loop in the context of a linear receive system (ie no IMD) of known Noise Figure. Continue reading Small untuned loop for receiving – simple model with transformer

## nanoVNA-H – Port 1 attenuator for improved what???

At nanoVNA-H – Port 2 attenuator for improved Return Loss I explained the reasons for essentially permanent attachment of a 10dB attenuator to Port 2 (Ch 1 in nanoVNA speak).

Above, the 10dB attenuator is semi permanently attached to Port 2 principally to improve the Return Loss (or impedance match) of Port 2, a parameter that becomes quite important when testing some types of networks than depend on proper termination (eg many filters). I should remind readers that the improvement in Port 2 Return Loss comes at a cost, the dynamic range of Port 2 is reduced by 10dB. Continue reading nanoVNA-H – Port 1 attenuator for improved what???

## Return Loss Bridge – some important details

Articles describing how to make a Return Loss Bridge are pretty common, but they don’t often spell out component values that are critical to accuracy.

Above is a schematic for discussion. It is somewhat simplified, but it is complete and will work. Continue reading Return Loss Bridge – some important details

## nanoVNA – promotion by cheats

A friend wrote saying “I thought the nanoVNA display was smaller than this”.

I make the index finger nail width exactly the same as the round part of the SMA nut which is 7.6mm. That is a very tiny hand… or the image is a composite fraudulently not to scale. Continue reading nanoVNA – promotion by cheats