One often sees newbies ask about their VSWR meter readings, and a common observation is that the measured VSWR is better at low power and as power is increased, VSWR increases.
With the evolution of the ‘shack in a box’, and knowledge and experience to match, the problem is often reported observed with the transceiver’s internal VSWR meter.
Some of these ‘shack in a box’ have some pretty nifty features, for example the very popular Icom IC-7300 not only has an internal VSWR meter for the HF bands, but it can perform an assisted sweep and display the results graphically.
Isn’t that a great idea, so convenient, all good!
Or is it? Continue reading Should you trust your VSWR meter?
From time to time one sees discussion online about consistency of ‘measured’ VSWR at different power levels (on the same instrument).
A question often asked is:
I tune up at 10W and achieve VSWR=1.5, and when I increase power to 100W, the VSWR increases. Which should I believe?
The first thing to note is that good antenna systems SHOULD be linear, VSWR should be independent of power, it is if the system IS linear.
For the most part they are linear, even though many antenna systems contain elements such as ferrite cored inductors that may exhibit some small level of non-linearity in ‘normal’ operation.
Non-linearity caused by for instance saturation of magnetic materials, loss of permeability where the temperature of ferrite cores reaches Curie point, arcing of capacitors or other insulating materials is NOT normal linear operation of a GOOD antenna system. If high indicated VSWR at high power is caused by any of these effects, it is flagging a problem that requires attention.
That said, a significant non-linear element may be the VSWR meter itself.
A common, if not most common way to make these meters is to use a half wave detector to convert the direction coupler RF outputs into DC to drive an ordinary moving coil meter. These meters commonly assume that the detector DC output voltage is exactly proportional to the RF input voltage.
Lets look at the accuracy of that process.
Above is a plot of the detector output vs RF input voltage for a commercial 200W VSWR meter. The measurements cover input power from 10 to 100W.
Continue reading VSWR meter trap for the unwary
This article reports initial impressions of an Anytone AT-D868UV hand held VHF/UHF dual mode (DMR/FM) radio.
Above, the AT-D868UV, purchased for about A$225 incl post from Hong Kong. This model had a GPS though that is unusable on ham DMR networks, so it is wasted money if you like. They may be more expensive through online shops that collect GST, and of course in countries where tariffs are applied to make them great again, prices may be higher.
Continue reading Anytone AT-D868UV: initial impressions
The WSPRlite flex requires external low pass filters for each band of operation.
Since my experiments will be conducted on the 40m band, the following low pass filter meets the requirement. The inductors and capacitors make a seven element Chebyshev filter as designed by G3CWI for use in a 50Ω system.
Above, the filter is assembled on a piece of matrix board with two BNC connectors. The inductors are fixed with hot melt adhesive, and the whole thing served over with heatshrink tube. It is not waterproof. Continue reading 40m filter for WSPRlite flex
I bought a remote speaker-microphone (RSM) for a DMR portable from eBay (~$12 posted). Experience says that these suffer RF ingress which is seriously bad in DMR due to the amplitude modulation of the transmitted signal.
This RSM had somewhat improved filtering around the electret compared to others I have purchased. Continue reading Another RFI mod of a speaker mic for DMR use
A correspondent wrote asking about the design of a matching network for a Half Square antenna for 80m, voltage fed at one end.
Above is the current distribution on the half square voltage fed. It is essentially two in-phase vertical quarter waves separated a half wavelength, a broadside array.
Feed point impedance at resonance is very high 5700Ω, and being a high Q antenna, they are very sensitive to dimensions, nearby clutter etc. Note that this is calculated for an antenna in the clear, it will be different where trees or conductive mast exist nearby. Continue reading 80m voltage fed Half Square matching workup
At Surecom SW-102 VSWR meter review I wrote a review of a meter which I had purchased a little over a year ago, it was at v4.5.
One of the many problems identified was inconsistency of displayed values.
Surecom’s versions are confusing, the highest number is not necessarily the latest version. It appears a partial version history from their current page advertising the SW-102 is:
OLD VERSION : V3.3 ,V3.8 ,V4.5,V4.9 ,V5.0,V5.1
2017-8 NEW VERSION : V2.02 ,V2.03
The following image is from Surecom’s current page advertising the SW-102, and I assume that the version shown here (v2.6) is the latest at time of writing.
The image captures the outputs of two tests with poor and good dummy loads.
Let’s check the displayed values for internal consistency. Continue reading Surecom SW-102 VSWR meter review – v2.6
This article describes a prototype RF choke (RFC) for use in a power injector for 50Ω coax over range 1.8-30MHz. Power injector / extractors are often used to connect power and / or signalling on a shared common RF coax feed line to accessories such as remote antenna switches and ATUs.
Design criteria are:
- Insertion VSWR of the RFC in shunt with 50+j0Ω < 1.1;
- Dissipation < 2% of a 100W transmitter.
The core chose is a LF1260 ferrite suppression bead from Jaycar. It is a medium / high µ core readily available in Australia at $7.50 / 6.
Above is the prototype RFC wound with data cable wire for the purpose of measurement. In application it could be wound with 1mm enamelled copper or PTFE insulated wire (Curie point is lowish at 120°+, but it still benefits from higher temperature insulation). Continue reading An RF choke for a 1.8-30MHz coax power injector – LF1260 core
In a discussion about using a 40m centre fed half wave dipole on 80m, the matter of feed line loss came up and online expert KM1H offered:
Use this to help make up your mind. Add it to the normal coax loss. http://www.csgnetwork.com/vswrlosscalc.html
This is to suggest that the feed line loss under standing waves can be calculated with that calculator.
He then berates and demeans a participant for commenting on his recommendation, bluster is par for the course in these venues.
The calculator in question states
this calculator is designed to give the efficiency loss of a given antenna, based on the input of VSWR (voltage standing wave ratio) and other subsequent factors.
This is a bit wishy washy,
efficiency loss is not very clear. The usual meaning of efficiency is PowerOut/PowerIn, and the usual meaning of loss is PowerIn/PowerOut, both can be expresssed in dB: LossdB=10*log(Loss) and EfficiencydB=10*log(Efficiency). Continue reading Line loss under standing waves – recommendation of dodgy tool on eHam
Richard (G3CWI) published an interesting blog article Comparison of groundwave performance of Small Transmitting Loop and Quarterwave GP summarising a recent WSPR test on 40m over 20km distance.
This article is a walk through of the expected WSPR receive S/N for the case of the 20mW tx on a quarter wave vertical.
100% efficient tx and rx antenna systems
Ground wave suffers attenuation due to two key components:
- dispersion of energy as the wave spreads out from the source; and
- absorption of energy in heating the soil.
Item (1) is simply inverse square law effect, and Norton provides us with several approximations for estimating (2) from Sommerfields work.
Calculate efficiency of vertically polarised antenna from far field strength uses Norton’s f5 approximation for ground wave attenuation.
Above is a calculation for a 100% efficient transmitter. (The trick to getting this is to leave the measured field strength field empty and the calculator will insert the value that gives 100% efficiency.)
So the next question is what ambient noise level might we expect in a rural setting on 40m. Continue reading G3CWI’s ground wave tests Jul 2017 using WSPRlite