One of the many nanoVNA cloners makes an interesting little inexpensive demo board with a selection of components, filters etc to develop familiarity with the nanovna.
Above is a pic of the demo board and the supplied jumper cables. The demo board may not include information relevant to using the cables and connectors supplied. Continue reading nanoVNA – that demo board and its U.FL connectors
At https://groups.io/g/nanovna-users/message/9185 a user posted a measurement made with his nanoVNA of a length of coax with termination.
Above is his initial reported measurement of
an approximate 350′ length of coax with a known good dummy load on the opposite end. 350′ is 106.7m. Whilst this chart is less value than a Smith chart rendering, understanding the nature of things allows us to infer the Smith chart. Continue reading nanoVNA user post provides an interesting example for study #1
The article Antennas – disturbing the thing being measured – open wire lines #5 demonstrated an inconsistency between the notion of a balun Common Mode Rejection Ratio CMRR property and a complete NEC model for predicting common mode current behavior.
In that case, two scenarios were modelled with only a change in the feed line length, yet they showed quite different currents near the same balun.
A common metric bandied around is the Common Mode Rejection Ratio (CMRR) and the definition is a bit rubbery, but it tends to come down to the ratio of the magnitude of common mode current to the magnitude of differential current in a test scenario (usually a lab workbench… with intention that the metric is then applicable more generally). (Anaren 2005) gives a popular explanation.
It is worth noting that the conventional meaning of CMRR in relation to op amps is that it is the ratio of differential gain to common mode gain and large +ve dB values are goodness, and it makes sense. Common use in terms of baluns is the opposite, Anaren gives the expression CMRR=S1c/S1d which will give large -ve dB values as goodness, and which seems inconsistent with the descriptive name where a large ratio (more +ve dB value) would be goodness. The balun ‘crowd’s’ use of a -ve rejection ratio seems a bit tautological, as if they haven’t really thought this through, it is a bit like the hammy thing of talking about the attenuation of a length of coax as -xdB.
I don’t think CMRR is a useful property of baluns per se, certainly not as a component of practical antenna systems, so I have written this article to report common mode ratio (CMR) (being the ratio of common mode to differential mode current at the point of interest). CMR is not a property of the balun, it expresses the relationship between the magnitudes of the components of current at a point of interest.
Keeping in mind that the differential current and common mode current distributions are usually both standing waves in the general case (usually with different phase wavelength and therefore relative phase), another dimension of the antenna problem is to look at the current distribution on the feedline of the NEC model scenario used for this series of articles. The model used here is the 20m feed line height and current balun with Zcm=1130+j1657Ω.
Above is a plot of |Ic|, |Id| and CMR in the NEC-4.2 model scenario. Segments are numbered from the lower end to upper end of the 20m long feed line. Continue reading Antennas – disturbing the thing being measured – open wire lines #6
The articles Antennas – disturbing the thing being measured – open wire lines #3 Antennas – disturbing the thing being measured – open wire lines #4 demonstrated an inconsistency between a partial linear circuit model and a complete NEC model for predicting common mode current behaviour.
One of the oft proposed solutions to characterising a balun is to find the Common Mode Rejection Ratio (a term carried over from other applications, eg voltage driven operational amplifiers). (Anaren 2005) explains a method of finding balun CMRR. (Skelton 2010) goes so far as to say
The Common Mode Rejection Ratio (CMRR) of a balun is defined in professional literature as the ratio of wanted to un-wanted transmitted power. As rejection of common-mode transmission is the primary purpose of a balun, it follows that CMRR should be the key figure of merit.
Let us take the model scenario used in Antennas – disturbing the thing being measured – open wire lines #4 and lower the height of the dipole 10 10m, and compare the model ratio of Ic/Id.
Again, we will use Python to do the complex maths for the without and with scenarios at 20m height, and without and with at 10m height. Continue reading Antennas – disturbing the thing being measured – open wire lines #5
The article Antennas – disturbing the thing being measured – open wire lines #2 did a simple analysis of current flows in the model scenario using ideal voltage balun drive with a current balun.
It was mentioned that solution of the lumped values network is only a first approximation and not as good as the NEC solution which properly models the coupled conductors, and their mutual effect on the distribution of currents in the system.
This article reports the NEC current solution decomposed into differential mode and common mode components.
Ideal voltage balun drive
Above is a zoomed in view of the feed point with a balanced pair of voltage sources feeding the line against the ground electrode. Continue reading Antennas – disturbing the thing being measured – open wire lines #4
The article Antennas – disturbing the thing being measured – open wire lines #2 did a simple analysis of current flows in the model scenario using ideal voltage balun drive.
That begs the question, what difference would a good current balun make?
We can get a good approximation of what happens by inserting the current balun’s Zcm in series with Z3. Let’s take Zcm to be 1130+j1657Ω (11t on a FT240-43). Continue reading Antennas – disturbing the thing being measured – open wire lines #3
The article Antennas – disturbing the thing being measured – open wire lines illustrated the pitfalls of a simplistic model of an antenna presented on two terminals of an open wire line.
A more complete representation of the antenna can be formed by making three impedance measurements (Schmidt nd). Continue reading Antennas – disturbing the thing being measured – open wire lines #2
A common question in online forums relates to inability to reconcile analyser measurements of an antenna system with the transmitter system antenna facing VSWR meter.
The cause is often that the antenna system was changed significantly to connect the analyser.
Seeing recent discussion by the online experts of how the measure the impedance of an antenna system looking into a so-called balanced feed line gives advice that is likely to cause reconciliation failure.
I will make the point firstly that the line is not intrinsically balanced, it is the way the it is used that may or may not achieve balance of some type. I will refer to that type of line as open wire line.
Let’s explore the subject using some NEC models.
I have constructed an NEC-4.2 model of an approximately half wave dipole at 7MHz, it is 20m above the ground, and fed slightly off centre with open wire line constructed using GW elements. At the bottom, I have connected a 2 segment wire between the feed line ends, and two sources in series. Continue reading Antennas – disturbing the thing being measured – open wire lines
An oft cited advantage of the nanoVNA are choices:
- hardware (several clones of the basic thing, the ‘improved’ -H series, the coming -H with bigger screen, the -F with bigger screen… and the future v2);
- firmware (lots and lots of forks, some hardware targeted);
- external clients (PC clients, web interfaces, Python / Octave / Matlab code etc).
There is not necessarily interoperatibilty between all instances of each level of this tree. For example, nanovna-F may not share firmware images with the original nanoVNA and its clones, and vice versa due to a different display protocol.
Some PC clients support features not implemented in all current firmware versions, eg screen capture. Continue reading nanoVNA – a surfit of choices
The NanoVNA is a new low cost community developed VNA with assembled units coming out of China for <$50.
The NanoVNA uses PCB end launch SMA connectors, and if one is tightened to anywhere near the SMA specification torque of 1Nm, the assembly is ‘soft’ as the board flexes… a warning that this may cause damage (track cracking or outright separation of the SMA connector).
If you have a bare board, you can counter this torque with a wrench applied across the flats of the female connector, but in my -H, it is fitted in a plastic case and the flats are not accessible. Continue reading nanoVNA-H – coax connectors