At On Witt’s calculation of Matched Line Loss from Return Loss I discussed the common but flawed thinking that Matched Line Loss (MLL) can be calculated as half of the Return Loss of either a S/C or O/C section of transmission line.

The article discusses Witt’s calculation (half the average of Return Loss for S/C and O/C conditions) and notes that it can be a good approximation where the actual Zo is very close to the Zo on which the Return Loss measurement is based, and that the line loss is low.

This article looks at a case study of a section of low loss nominally 75Ω line is measured on a 50Ω instrument to illustrate sensitivity to Zo error.

A 3.1m section of RG6 was measured with O/C then S/C termination using a 50Ω VNA, and HalfReturnLoss (HRL), |S11| and phase of S11 is plotted.

Above, the O/C termination.

Above, the S/C termination.

## MLL estimates

### Simple Half Return Loss rule

Plotted Half Return Loss is all over the place, and it is clear from the chart that estimating MLL from HRL will not give the expected response of MLL increasing roughly proportional to square root of frequency for this cable at these frequencies.

The method fails on this data.

### Simple half Return Loss rule – variation

A variation of the HRL rule is to take HRL for either an O/C section where phase of S11 is 180°, or for a S/C section where phase of S11 is 0°.

Marker 1 is at such a frequency, so lets consider that ‘improved’ algorithm.

HRL=0.24dB for the O/C case, specification MLL for the section is 0.09dB. Not a good estimate alone.

HRL=0.12dB for the S/C case, specification MLL for the section is 0.09dB. Not a good estimate alone.

So, these methods fail miserably on this data.

### Witt’s calculation

Taking the values at Marker 1 (19.2MHz), HRL for O/C is 0.24dB, and for S/C is 0.12dB.

Witt’s calculation would give MLL=(0.24+0.12)/2=0.18dB for the section. This is twice the expected MLL for this cable type (0.09dB) for the section.

Witt’s method fails miserably in this case.

(Duffy 2012) shows that Witt’s method can provide a reasonable approximation where the Zo error is very small (such as the difference between nominal and actual Zo of a practical line at HF and above).

## Conclusions

This case study shows that even a small difference between actual Zo of a line section and the Return Loss measurement basis can result in gross error in the estimated MLL.

Even though these methods are common, and recommended in manuals for many instruments, they demand competent use to achieve reasonable accuracy.

## References

- Bird Electronic Corporation. 2004. RF directional thruline wattmeter model 43 – instruction book. Cleveland: Bird Electronic Corporation. p14.
- Duffy, O. Jul 2012. Estimating Matched Line Loss from Return Loss.
- Jennings, L. Aug 2012. Quick coax test In QST Aug 2012.
- Witt, F. May 2005. In QEX. http://www.arrl.org/files/file/Technology/tis/info/pdf/0505qex044.pdf (accessed 01/01/2015).