Measure inductor using OSL calibration
At Measuring balun common mode impedance I showed a method of backing out the effects of a text fixture using the “subtract cable” facility of Antscope software with the Rigexpert AA-600.
Some analysers (including the AA-600) support OSL calibration of the instrument itself, and some support OSL calibration using the client software (Antscope in this case). This article demonstrates use of Antscope with OSL calibration to measure a small RF inductor which has similar characteristic to good Guanella 1:1 HF baluns.
The text fixture used for this demonstration is constructed on a SMA(F) PCB connector using some machined pin connector strip, and SMA(M)-SMA(M) and N(M)-SMA(F) adapters to connect to the AA-600.
Above is a pic of the test inductor in the test fixture with adapters. The test inductor 6 turns of 0.5mm PVC insulated wire wound on a BN-43-202 binocular balun core. Continue reading Exploiting your antenna analyser #16
In Improved cooling for the MFJ-949E I described a solution to a problem of demonstrated overheating of the ATU at rated power, indeed at a lot less than rated power.
Though I have never measured the ATR-30 temperature rise, and am probably unlikely to stress the 3kW rated ATU with a 100W transmitter, I have performed a similar cooling modification to the ATR-30.
Continue reading Improved cooling for the ATR-30
Measure MLL using the half ReturnLoss method – a spot test with a hand held analyser
At Exploiting your antenna analyser #14 I gave an explanation of the method of approximating MLL of a line section by taking the average half Return Loss with o/c and s/c terminations.
This article demonstrates the technique using the Rigexpert AA-600 analyser in hand held mode.
The task is to assess whether a section of RG58A/U coax has MLL at 3.5MHz similar to specification or not.
The specification loss of 10.13m of RG58A/U has MLL=0.29dB.
Above, the first test with an o/c termination. Return Loss is 0.4dB. Continue reading Exploiting your antenna analyser #15
Assessing the Q of a half wave dipole antenna system explained that Q can be a valuable indicator of antenna system health.
This article uses a recently published VSWR curve for a 15m half wave dipole antenna system as an example to demonstrate the technique.
The following graph is from a Sark100 style antenna analyser, and it is quite a poor start to diagnostics, but using it draws out what is desired for further analysis.
Above, the captured VSWR(50) sweep. Continue reading Assessing the Q of a half wave dipole antenna system – a real world example
Q can be a valuable indicator of antenna system health
The Q of an antenna can be a useful statistic in assessing whether it is operating as it should.
The Q of half wave dipole antenna system on HF depends to some extent on conductor size, its environment (height, type of ground, nearby structures and vegetation, and feed line / matching loss. Nevertheless, it should usually fall in the range of 10 to 13 for good wire dipoles, and if you measure a half wave dipole antenna system to have Q significantly outside that range, it is probably significantly less efficient than it should be. Continue reading Assessing the Q of a half wave dipole antenna system
This article documents a field strength survey of an M40-1 short helical vertical on 40m.
This test is more a feasibility study of the experimental method and apparatus than an absolute measure of the antenna.
The antenna under test is described at AUT – MobileOne M40-1 40m helical.
Field strength was measured using a small square untuned loop and VK3AQZ RF power meter (RFPM1), and data was captured using A prototype data logger for RFPM1.
VK3AQZ RF power meter (RFPM1) described my build and calibration of the RFPM1.
Above is the RFPM1, shown with two probes, but only one probe is required for this procedure, the other is disconnected. The RFPM1 directly reads input power in dBm.
The loop antenna used was described at (Duffy 2007). It is a small square loop (600mm sides) fed in one corner with a 1:1 voltage balun. Continue reading Field strength survey of an M40-1 short helical vertical on 40m
At Improved cooling for the MFJ-949E I described a modification to the ATU to improve its cooling using a fan and run on timer.
The run on timer described was based on a Chinese STC15F104E DIP8 8051 like microcontroller.
Because the programming tools for the STC chips work so poorly, and the lack of documentation of their protocol, there is no simple way to update only the calibration data in EEPROM. I have ported the algorithm to an ATTINY25 which doesn’t cost a lot more but had a much better development environment and a range of tools to allow EEPROM update without overwriting the FLASH image, and as well it will run my bootloader, ATB.
This article describes a generic run on timer based on an Atmel AVR chip, a ATTINY25 though the code will also run in ATTINY45 and ATTINY85.
The circuit is very simple, the DC output from the forward power detector is connected to the input pin which turns the BC548C transistor on at input voltage greater than about 0.7V. The high value of base resistor ensures very light loading of the forward power detector.
Continue reading A generic run on timer using an ATTINY25
Sixth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
This article documents measurement of impedance.
The antenna system is a G5RV with tuned feeders (9m of home made 450Ω open wire). The tuned feeders terminate on the balun described in this series, and it is located on the outside of the antenna feed entrance panel shown above. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #6
Fifth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
Installation / testing
The balun packaged in a non-conductive housing was designed to have minimal stray capacitance to ground to minimise common mode current with asymmetric loads.
Above, the balun is attached to the exterior side of the antenna feed entrance panel using a male to male N adapter, done up very tight. The feed line connections are liberally coated with marine grease to prevent ingress of water and oxygen, a measure to reduce corrosion. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #5
At A look at internal losses in a typical ATU I demonstrated that it is quite easy to raise the temperature of the coil in the MFJ-949E to an unsafe level, even with quite modest power.
The most heat sensitive component in this ATU is the coil, specifically the coil supports which are probably polystyrene, and the glass transition temperature of polystyrene is around 100°.
This article documents modification of my MFJ-949E to reduce the risk of damage under some operating conditions. Continue reading Improved cooling for the MFJ-949E