Optimising a G5RV with hybrid feed
(Varney 1958) described his G5RV antenna in two forms, one with tuned feeders, and the more popular form with hybrid feed consisting of a so-called matching section of open wire line and then an arbitrary length of lower Zo coax or twin to the transmitter.
(Duffy 2005) showed that the hybrid feed configuration is susceptible to high losses in the low Zo line as it is often longish, is relatively high loss line and operates with standing waves. Varney did offer two options for the low Zo line:
any length 72Ω twin or coax. Continue reading Exploiting your antenna analyser #17
Measure inductor using OSL calibration
At Measuring balun common mode impedance I showed a method of backing out the effects of a text fixture using the “subtract cable” facility of Antscope software with the Rigexpert AA-600.
Some analysers (including the AA-600) support OSL calibration of the instrument itself, and some support OSL calibration using the client software (Antscope in this case). This article demonstrates use of Antscope with OSL calibration to measure a small RF inductor which has similar characteristic to good Guanella 1:1 HF baluns.
The text fixture used for this demonstration is constructed on a SMA(F) PCB connector using some machined pin connector strip, and SMA(M)-SMA(M) and N(M)-SMA(F) adapters to connect to the AA-600.
Above is a pic of the test inductor in the test fixture with adapters. The test inductor 6 turns of 0.5mm PVC insulated wire wound on a BN43-202 binocular balun core. Continue reading Exploiting your antenna analyser #16
Measure MLL using the half ReturnLoss method – a spot test with a hand held analyser
At Exploiting your antenna analyser #14 I gave an explanation of the method of approximating MLL of a line section by taking the average half Return Loss with o/c and s/c terminations.
This article demonstrates the technique using the Rigexpert AA-600 analyser in hand held mode.
The task is to assess whether a section of RG58A/U coax has MLL at 3.5MHz similar to specification or not.
The specification loss of 10.13m of RG58A/U has MLL=0.29dB.
Above, the first test with an o/c termination. Return Loss is 0.4dB. Continue reading Exploiting your antenna analyser #15
Assessing the Q of a half wave dipole antenna system explained that Q can be a valuable indicator of antenna system health.
This article uses a recently published VSWR curve for a 15m half wave dipole antenna system as an example to demonstrate the technique.
The following graph is from a Sark100 style antenna analyser, and it is quite a poor start to diagnostics, but using it draws out what is desired for further analysis.
Above, the captured VSWR(50) sweep. Continue reading Assessing the Q of a half wave dipole antenna system – a real world example
Q can be a valuable indicator of antenna system health
The Q of an antenna can be a useful statistic in assessing whether it is operating as it should.
The Q of half wave dipole antenna system on HF depends to some extent on conductor size, its environment (height, type of ground, nearby structures and vegetation, and feed line / matching loss. Nevertheless, it should usually fall in the range of 10 to 13 for good wire dipoles, and if you measure a half wave dipole antenna system to have Q significantly outside that range, it is probably significantly less efficient than it should be. Continue reading Assessing the Q of a half wave dipole antenna system
This article documents a field strength survey of an M40-1 short helical vertical on 40m.
This test is more a feasibility study of the experimental method and apparatus than an absolute measure of the antenna.
The antenna under test is described at AUT – MobileOne M40-1 40m helical.
Field strength was measured using a small square untuned loop and VK3AQZ RF power meter (RFPM1), and data was captured using A prototype data logger for RFPM1.
VK3AQZ RF power meter (RFPM1) described my build and calibration of the RFPM1.
Above is the RFPM1, shown with two probes, but only one probe is required for this procedure, the other is disconnected. The RFPM1 directly reads input power in dBm.
The loop antenna used was described at (Duffy 2007). It is a small square loop (600mm sides) fed in one corner with a 1:1 voltage balun. Continue reading Field strength survey of an M40-1 short helical vertical on 40m
Sixth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
This article documents measurement of impedance.
The antenna system is a G5RV with tuned feeders (9m of home made 450Ω open wire). The tuned feeders terminate on the balun described in this series, and it is located on the outside of the antenna feed entrance panel shown above. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #6
Fifth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
Installation / testing
The balun packaged in a non-conductive housing was designed to have minimal stray capacitance to ground to minimise common mode current with asymmetric loads.
Above, the balun is attached to the exterior side of the antenna feed entrance panel using a male to male N adapter, done up very tight. The feed line connections are liberally coated with marine grease to prevent ingress of water and oxygen, a measure to reduce corrosion. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #5
On review of the Ultrafire XML-T6 torch, I found the mode switching / mode memory so dysfunctional that it rendered the torch useless in my evaluation.
This article describes a work around that makes the thing usable (IMHO). Continue reading Ultrafire XML-T6 LED torch – a fix for the dysfunctional mode memory ‘feature’
The operating temperatures of refrigerators and freezers used for food storage is important to safe storage of food and to minimisation of energy costs.
The US FDA recommends the refrigerator should be set to 40F (4.4°) and the freezer to 0F (-17.8°).
Temperatures vary inside the cabinets, and they vary over time with opening and closing doors, and introduction of warmer goods for storage.
Many spot temperature checks are helpful but they don’t provide a very complete picture, and opening the door to make measurements disturbs the very thing being measured. Continue reading Fridge / freezer setup