A correspondent wrote seeking clarification of the Telepost LP-100A claims re impedance measurement in the context of some of my previous articles on the sign of reactance.

I could see several mentions in the LP-100A manual and the LP_100Plot documentation and they do seem a little inconsistent.

The LP-100A manual states very clearly:

Note: The LP-100A cannot determine the sign of X automatically.

and;

If you QSY up from your current frequency, and the reactance goes up, then the reactance is inductive (sign is “+”), and conversely if it goes down, then the reactance is capacitive (sign is “-“). A suitable distance is QSY is about 100 kHz or more. The LP-Plot program has the ability to determine sign automatically, since it can control your transmitter’s frequency. When it plots a range of frequencies, it uses the slope of the reactance curve to determine sign, and plots the results accordingly.

The first part states clearly that the instrument cannot directly measure the sign of reactance, and presumably measures the magnitude of reactance |X|.

Lets explore the second part in light of the overarching statement of the first part.

Above is the calculated R and X looking into 7m of Belden RG58C/U with a load 25+j0Ω. Also shown is |X|(as would be measured by the LP-100A) and calculated magnitude of phase of R,X, |φ|. Continue reading LP-100A impedance measurement

Last update: 4th January, 2017, 10:31 AM