Insertion Loss, Mismatch Loss, Transmission Loss
A correspondent having read Exploiting your antenna analyser #12 asks whether the measurement provides evidence of loss of the connectors, and referred me to (Arther nd) where he reports some measurements of UHF series adapters and conclusions.
Let’s deal with interpretations of my own measurements first.
Measurements of input impedance only for such an electrical short transmission line will not give useful data for determining TransmissionLoss which is the result of conversion of RF energy to heat. The measurements do give ReturnLoss and given that InsertionLoss=MismatchLoss+TransmissionLoss, they set a lower bound for InsertionLoss.
Above is a plot of ρ and ReturnLoss for the DUT. ReturnLoss curiously is plotted ‘upside down’ as ReturnLoss increases downwards… a quirk of AIM software, but remember that ReturnLoss in dB is +ve.
Continue reading Exploiting your antenna analyser #13