## Resolving the sign of reactance – a method

Many analysers do not measure the sign of reactance, and display the magnitude of reactance, and likewise for magnitude of phase and magnitude of impedance… though they are often incorrectly and misleadingly labelled otherwise.

The article The sign of reactance explains the problem and dismisses common recipes for resolving the sign of reactance as not general and not reliable.

This article gives an example of one method that may be useful for resolving the sign of reactance.

My correspondent has measured VSWR=1.68 and |Z|=66 and needs to know R and X. From those values we can calculate R=60.3 and |X|=26.9.

## Method

The method involves adding a short series section of known line, short enough to provide a measurement difference in R, and that R would be different for the case of =ve and -ve X, all of these measured at the same frequency. Continue reading Exploiting your antenna analyser #28

Last update: 12th July, 2017, 11:33 AM