## Finding resistance and reactance with some low end analysers #2

Exploiting your antenna analyser #8 was about finding resistance and reactance with some low end analysers that don’t directly display those values of interest. The article showed how to calculate the values starting with |Z| from the analyser and included links to a calculator to perform the calcs.

This article describes an extension to that calculator Find |Z|,R,|X| from VSWR,|Z|,R,Ro to use R, VSWR, and Ro as the starting point. Note that the sign of X and the sign of the phase of Z cannot be determined from this starting point, there just isn’t enough information.

You will probably not find the equation for |X|(R,VSWR,Ro) in text books or handbooks, and the derivation is not shown here but if there is interest, I may publish a separate paper.

Lets say you knew VSWR=2, R=75Ω, Ro=50Ω, what is |X|?

Above, entering the values in the calculator we find that |X|=35.4Ω. Continue reading Exploiting your antenna analyser #20