This article builds on Accuracy of AIMuhf system – AIM865A vs AIM882 on a ferrite cored inductor by adding a measurement using AIM885 of the same test inductor.
Above is the test inductor. It comprises 6 turns of 0.5mm PVC insulated wire wound on a BN-43-202 binocular balun core. Continue reading Accuracy of AIMuhf system – AIM865A vs AIM882 vs AIM885 on a ferrite cored inductor
New features in version 885 is
Fixed “Refer to Antenna” function… so lets check it out.
The AIMuhf was calibrated using the supplied OSL components and AIM885.
Above is a sweep of a 1.05m length of Belden 8262 (RG58C/U) cable (with o/c termination) attached to the AIMuhf. Continue reading AIM885 Refer To Antenna checkout
In other posts, I have commented on the apparent inconsistency of AIMuhf measurements.
One of the devices I often wish to measure is a ferrite cored choke such as those used for a Guanella 1:1 balun. A small test inductor was made to provide a common device for measurement across instruments, and versions of software, though small, it has similar characteristics to the larger inductors more commonly measured.
Above is the test inductor. It comprises 6 turns of 0.5mm PVC insulated wire wound on a BN-43-202 binocular balun core. Continue reading Accuracy of AIMuhf system – AIM865A vs AIM882 on a ferrite cored inductor
At Where is the best place to measure feed point VSWR I discussed location of the VSWR meter and projection of its reading to another point on a known transmission line.
A correspondent has taken me to task and citing Telepost’s LP-100A manual: Continue reading LP-100A manual advice on VSWR measurement
Jaycar have sold the MP3090 (Manson SPS-9400) over many years. I have one dated 2001 and they are still in their catalog in 2015. Manson is a Chinese manufacturer of medium quality power supplies, at reasonable prices. Continue reading Jaycar MP3090 / Manson SPS-9400 repair
At Diamond X-50N #2 at VK2OMD I reported some details of a new antenna at VK2OMD, including some baseline measurements using a Rigexpert AA-600.
This article expands that information with some measurements made using AIMuhf, and gives some contrasts between the two tools.
The AA-600 measurements were made 26/11/2014, and AIMuhf a couple of months later on 29/01/2015. The system measured should have been identical, the feed line is 9.83m of LDF4-50A with N-type connectors at both ends. Continue reading Diamond X50N measurements – AIMuhf and AA-600
I have spent considerable time over the last few weeks trying to salvage something useful and reliable from the wreckage that is AIMuhf and its PC client program AIM. Continue reading AIMuhf – my strategy
I have had cause to validate the output produced by an AIMuhf measurement using AIM882 (current version, released about three months ago).
The test scenario is a pair of nominal 50+j0Ω loads on a Tee piece, connected to the AIMuhf by about 1m of RG58 coax and swept from 10 to 50MHz.
It is mental arithmetic that the VSWR should be very close to 2:1, and since the loss of the cable is quite low, VSWR should be almost uniform with frequency. Continue reading AIM 882 produces internally inconsistent results
There is often a need for a 9V battery for portable test equipment (NNA, Noise Bridge, Low R meter, Power Meter etc). A solution is a 8 cell NiMH pack.
Above, a battery pack made from two Hobbyking 4 low self discharge AA cells. The packs come with JR servo connectors, and the pins are rewired to use the -ve from one pack and +ve from the other pack to one of the JR connectors. The other wires are connected via a 3A Polyswitch for s/c protection. A short JR to 2.1mm DC connector is made from a JR extension cable and 2.1mm connector. Continue reading Inexpensive utility rechargeable 9V battery pack for test instruments