In Improved cooling for the MFJ-949E I described a solution to a problem of demonstrated overheating of the ATU at rated power, indeed at a lot less than rated power.
Though I have never measured the ATR-30 temperature rise, and am probably unlikely to stress the 3kW rated ATU with a 100W transmitter, I have performed a similar cooling modification to the ATR-30.
Continue reading Improved cooling for the ATR-30
Measure MLL using the half ReturnLoss method – a spot test with a hand held analyser
At Exploiting your antenna analyser #14 I gave an explanation of the method of approximating MLL of a line section by taking the average half Return Loss with o/c and s/c terminations.
This article demonstrates the technique using the Rigexpert AA-600 analyser in hand held mode.
The task is to assess whether a section of RG58A/U coax has MLL at 3.5MHz similar to specification or not.
The specification loss of 10.13m of RG58A/U has MLL=0.29dB.
Above, the first test with an o/c termination. Return Loss is 0.4dB. Continue reading Exploiting your antenna analyser #15
At Improved cooling for the MFJ-949E I described a modification to the ATU to improve its cooling using a fan and run on timer.
The run on timer described was based on a Chinese STC15F104E DIP8 8051 like microcontroller.
Because the programming tools for the STC chips work so poorly, and the lack of documentation of their protocol, there is no simple way to update only the calibration data in EEPROM. I have ported the algorithm to an ATTINY25 which doesn’t cost a lot more but had a much better development environment and a range of tools to allow EEPROM update without overwriting the FLASH image, and as well it will run my bootloader, ATB.
This article describes a generic run on timer based on an Atmel AVR chip, a ATTINY25 though the code will also run in ATTINY45 and ATTINY85.
The circuit is very simple, the DC output from the forward power detector is connected to the input pin which turns the BC548C transistor on at input voltage greater than about 0.7V. The high value of base resistor ensures very light loading of the forward power detector.
Continue reading A generic run on timer using an ATTINY25
Sixth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
This article documents measurement of impedance.
The antenna system is a G5RV with tuned feeders (9m of home made 450Ω open wire). The tuned feeders terminate on the balun described in this series, and it is located on the outside of the antenna feed entrance panel shown above. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #6
Fifth part in the series documenting the design and build of a Guanella 1:1 (current) balun for use on HF with wire antennas and an ATU.
Installation / testing
The balun packaged in a non-conductive housing was designed to have minimal stray capacitance to ground to minimise common mode current with asymmetric loads.
Above, the balun is attached to the exterior side of the antenna feed entrance panel using a male to male N adapter, done up very tight. The feed line connections are liberally coated with marine grease to prevent ingress of water and oxygen, a measure to reduce corrosion. Continue reading Design / build project: Guanella 1:1 ‘tuner balun for HF’ – #5
At A look at internal losses in a typical ATU I demonstrated that it is quite easy to raise the temperature of the coil in the MFJ-949E to an unsafe level, even with quite modest power.
The most heat sensitive component in this ATU is the coil, specifically the coil supports which are probably polystyrene, and the glass transition temperature of polystyrene is around 100°.
This article documents modification of my MFJ-949E to reduce the risk of damage under some operating conditions. Continue reading Improved cooling for the MFJ-949E
The so-called End fed Zepp (EFZ) is often cited as the basis for many more recent antenna designs, and is leveraged to provide and explanation… though few hams understand how the EFZ actually works.
End fed Zepp
Above is a diagram from the ARRL Antenna Handbook (Silver 2011).
Though a short conductor is shown to the right of the right hand feed line wire, the length is not specified or discussed in the accompanying text. It is popularly held that this is a “counterpoise” that provides a path for current equal to that flowing left into the main horizontal wire.
Let us explore the EFZ using NEC. The models are a reflection left to right of the above diagram, ie the feed is on the left hand end. Continue reading End fed Zepp
There seems to have been a revival in use of the so-called End Fed Half Wave antenna.
The prospect that a small radio such as the FT817, a magic match box and 10m of wire makes a good 20m field station appeals to many a SOTA enthusiast.
Let us model a scenario with a FT817 powered by internal battery and sitting on an insulating platform (eg a pack) 0.3m above natural ground, a 10m wire strung up into a tree at a 45° angle, and a 1m long mic cord stretched up at 45° in the other direction. The is the popular so-called ‘no counterpoise’ configuration.
A simplified model of just the current paths without regard to the bulk of the radio, or the effect of the helix of the mic cord illustrates an approximate current distribution. The model uses 1W RF input to the antenna over ‘average ground’ (0.005,13).
Above is a plot of the current distribution. Current is a minimum at the open ends, a boundary condition for the problem, and maximum in the middle of the half wave. We expect H field to be greatest near the current maximum, and E field to be greatest near the current minima. Continue reading Thinking about SOTA, EFHW and EMR safety
RG-6 has become a popular 75Ω transmission line for ham stations, and I have used it to good effect in many applications.
(Duffy 2007) extolled the virtues and gave implementation information, but cautioned:
Some types of RG−6/U use a CCS centre conductor and will have higher loss at low frequencies that shown in Fig 1, depending on the thickness of the copper cladding which may vary from cable to cable.
I have used RG-6/U with solid copper centre conductor widely on HF, and measured performance has always been consistent with expectation.
However, RG-6/U with solid copper centre conductor has become very hard to obtain, and products that remain available such as Belden 1694A are quite expensive.
This article documents measurements at low HF on a 100m roll of Quad shield RG-6/U purchased for UHF TV cabling.
The method used was to measure input impedance of the open circuit terminated 100m line section at a range of resonant and antiresonant frequencies, and from those to calculate Matched Line Loss (MLL) in dB/m.
Above is an example measurement around 3.74MHz. Zin is 213.4Ω at 3.74MHz. In this case I have used an AIMuhf one port analyser, but any instrument that can measure impedance in the range 10-500Ω would suit this particular scenario. Measurement of short low loss cables will yield more extreme impedances and may not be within range of some instruments. Continue reading RG-6/U for lower HF
Eric posed a quick quiz for the masses to test their knowledge under his heading “Do you really understand impedance matching?”
For your convenience, I will quote his challenge here.
All connections are made with low-loss coaxial cable. The antenna tuner is high quality with negligible losses.
According to standard conversion charts, we find that 4:1 SWR will give us 36% reflected power. Keep that number handy.
Now, we set up the experiment. First, set the slugs on BOTH wattmeters to read REFLECTED power.
Turn on the transmitter, and adjust the antenna tuner for zero reflected power on Bird #1. Switch to forward power, and set transmitter output to exactly 100 watts. Readjust antenna tuner if necessary to achieve zero reflected power, while maintaining 100 watts forward.
Go to Bird #2 and confirm that reflected power is 36 watts.
Question: What is the FORWARD power on Bird Wattmeter #2? How you answer this question determines if you understand the conjugate match theorem or not.
Let us assume that the transmission lines are 50Ω, and that the Bird wattmeters are calibrated for 50Ω.
So, to extract the key information, we have a lossless system (KL7AJ is a lossless kind of guy) and the load is stated to be VSWR=4. Continue reading KL7AJ quick quiz 21/02/2016